The 13519 is a 3.6 Vdc, 1.2 Ahr, 1/2 AA size cell battery.
The 13519 fits inside a CR800, CR850, CR1000, or CR3000 datalogger. This lithium battery powers the clock and SRAM when the primary 12 Vdc power supply is not connected. The 13519 is also used in many of the Campbell Scientific water samplers.
The 13519 needs to be periodically replaced. Users can replace the battery by following the process illustrated in the product manual. Typically, the battery is replaced when its voltage is at or below 2.7 Vdc.
|Self-Discharge Rate||1%/year (@ 20°C)|
|Temperature Range||-55° to +85°C|
Number of FAQs related to 13519: 1
The internal battery is not rechargeable. In the case of low internal battery voltage, data will only be maintained as long as a power supply is connected. Collect all data from the data logger before disconnecting the power. Then, replace the internal battery.