Frequently Asked Questions about Our Products and Solutions
Ask a Question
Ask a Question
(1828 total: showing 1-20)
Does the CR1000X have a Thermistor() instruction similar to the CR6?
In the event that the CR1000X becomes damaged and needs a new wiring panel, what should be done?
How does the power usage of the CR1000X compare to the power usage of the CR1000 or CR6?
IF PC208W is running on Windows XP, can it be set up to communicate with a CR1000X?
Can a program written for a CR1000 be used on a CR1000X?
After the release of the CR1000X, will the CR1000 continue to be sold and supported?
How can the current OS version of a CRBasic datalogger be viewed?
Can a CS320 be calibrated to a smaller range to measure only PAR (photosynthetically active radiation)?
How can the CS320 be mounted if the 18356 leveling base is not used?
Is it possible to read the output from a CS320 in a 4 to 20 mA form?
On routine site visits, how can the calibration of the CS320 be checked?
How often does the CS320 need to be recalibrated?
How long can the CS320 cable be?
What is the difference between a thermopile pyranometer and a silicon photocell pyranometer?
How can an end user simulate a lightning strike to an SG000 or CS110?
How many AM16/32B multiplexers can be connected to a datalogger at the same time?
How many AM25T multiplexers can be connected to a CR6 datalogger?
Which weather measurements are needed when calculating reference evapotranspiration?
If a CH200 or PS200 is hooked up to both solar and ac power supplies, which power supply does the device use?
If a CH150 or PS150 is hooked up to both solar and ac power supplies, which power supply does the device use?