Frequently Asked Questions about Our Products and Solutions
Ask a Question
Ask a Question
(1835 total: showing 1-20)
What is the difference between LoggerLink and LoggerNet Mobile Connect?
To prepare the molecular sieve, how many grams are needed per bottle?
Which countries can the molecular sieve be shipped to?
Why did Campbell Scientific start using molecular sieve? Does it have the same effect as the previously used chemicals?
Can molecular sieve be used in the future to replace bottles for gas analyzers sold before July 2017?
What might cause a battery to not hold its charge?
If a CS110 electric field sensor is installed in complex terrain, such as on a ridgeline in alpine territory, what is the effective range (spatial range) of the instrument, and how might it differ from a flatland site?
Does the CR1000X have a Thermistor() instruction similar to the CR6?
In the event that the CR1000X becomes damaged and needs a new wiring panel, what should be done?
How does the power usage of the CR1000X compare to the power usage of the CR1000 or CR6?
IF PC208W is running on Windows XP, can it be set up to communicate with a CR1000X?
Can a program written for a CR1000 be used on a CR1000X?
After the release of the CR1000X, will the CR1000 continue to be sold and supported?
How can the current OS version of a CRBasic datalogger be viewed?
Can a CS320 be calibrated to a smaller range to measure only PAR (photosynthetically active radiation)?
How can the CS320 be mounted if the 18356 leveling base is not used?
Is it possible to read the output from a CS320 in a 4 to 20 mA form?
On routine site visits, how can the calibration of the CS320 be checked?
How often does the CS320 need to be recalibrated?
How long can the CS320 cable be?