Frequently Asked Questions about Our Products and Solutions
Ask a Question
Ask a Question
(1911 total: showing 1-20)
Is the calibration for the CS240DM NIST traceable?
Are the traditional methods of using RTD to measure the back-of-module temperature not used with the CS240 and CS240DM because these methods may cause hot spots?
Can multiple instances of Campbell Scientific SURVEYOR be run on a PC at the same time?
How tight do the wing-nuts on the CM226 integrated mounting stand need to be tightened to secure the solar sensor?
Can the SG000 determine the exact distance to a lightning strike?
How do I cancel a subscription for cellular data services?
Can I suspend my subscription for cellular data services?
Can I order more than a one-year subscription of cellular data services?
Can a cellular data services subscription be transferred to another customer?
How do I renew my cellular data services subscription?
What happens if my data use goes over the monthly allowance for my cellular data services plan size?
If I have a cellular data services plan, where can I find modem status information?
When selecting a recalibration service for a pyranometer, how do the services differ and what is the importance of ISO 9060?
What are some distinguishing features of the MetSENS compact weather sensors?
Can the CSAT3BH heaters be controlled manually?
What happens to the data when the CSAT3BH heaters are on?
Is there a version of LNDB for Linux?
How often should the CR1000 be recalibrated?
When should a data acquisition (DAQ) system be used instead of a PC-based system?
How can the age of a sealed rechargeable battery be determined?