Frequently Asked Questions about Our Products and Solutions
Ask a Question
Ask a Question
(1932 total: showing 1-20)
Why are single-ended measurements used in the
instruction in CRBasic when the outputs of the potential transformer and current transformer are inherently differential?
Does the CS241 have a protection rating?
Does the mast of the CM375 have any lightning protection?
If duckbill anchors cannot be used to stabilize the CM375 in uneven, rocky terrain, what should be used instead?
Can the base plate of the CM375 be fixed to the ground to improve stability, especially in rocky, mountainous areas?
Can the mast and the guy cables of the CM375 support ice loading?
Under what temperature conditions can the CM375 operate?
What is the probable life expectancy of the CM375 mast?
Is the CS241 recommended for use at the back or the front of the bifacial module?
For the standard (non-NIST) version of the CS240 sensors, how does Campbell Scientific verify the ability of the sensors to measure and that the measurements made are calibrated to provide correct measurements?
How can data from LoggerNet be added into the SCADA software OSIsoft PI?
Why is NIST traceability beneficial for surface temperature sensors, such as the CS240, the 110PV, and the CS240DM?
If the temperature drops below -40°C, is that a problem for the CSAT3BH?
How does the CH201 compare with the 18529 Morningstar SunSaver-10 Regulator?
Can the CH201 be connected to two 10 A sources, or does the total from the two sources have to be equivalent to 10 A?
If a CH201 is connected to both a solar panel and an AC/DC power convert, which power supply does the device use?
How quickly can measurements be attained using the ClimaVUE™50?
Is the calibration for the CS240DM NIST traceable?
Are the traditional methods of using RTD to measure the back-of-module temperature not used with the CS240 and CS240DM because these methods may cause hot spots?
Can multiple instances of Campbell Scientific SURVEYOR be run on a PC at the same time?