The BaroVue™10 is a highly accurate barometer that can measure pressures over a range of 500 to 1100 hPa and can be used in a range of applications that includes meteorology and hydrology. This sensor includes our zero downtime recalibration. Other sensors on the market need to be taken offline and sent back to the manufacturer to be recalibrated, often for weeks at a time, causing a loss in valuable data and time, as well as an additional expense.
The sensor card for the BaroVue 10 is pre-calibrated. Replacement sensor cards are also pre-calibrated, requiring only one trip to the field station and no downtime. The design of the BaroVue 10 allows you to remove and replace your sensor card quickly and easily without either having to disassemble the equipment in the enclosure or having to send the equipment back to the manufacturer.Read More
The BaroVue 10 is a low-power, digital barometer (-40 to +60°C) that provides best-in-class accuracy and stability (±0.1 hPa/yr) over the entire pressure and temperature range. It can run continuously and does not require power cycling unlike other products in this category. The digital output is both SDI-12 and RS-232, which reduces noise and measurement uncertainty compared with analog sensors, and makes the BaroVue 10 compatible with all Campbell Scientific SDI-12 devices. Moreover, this barometric pressure sensor can be installed and integrated into many new and existing weather stations. The transducers used in the BaroVue 10 are direct-to-digital sensors, and no reconversion takes place inside the barometer.
|These specifications assume the recommended desiccation is used.
|500 to 1100 hPa
|-40° to +60°C
|9 to 28 Vdc
|~609.6 m (2,000 ft) below sea level (as in a mine) to 4,572 m (15,000 ft) above sea level
|SDI-12, RS-232 serial
|Barbed fitting for 0.318 cm (0.125 in.)
|2.2 x 9.0 x 10.2 cm (0.87 x 3.54 x 4.02 in.)
|226.8 g (0.5 lb)
|±0.3 hPa (at 20°C)
|±0.5 hPa (at -40° to +60°C)
|Power Supply Rejection
|0.05 hPa (RMS)
Note: The following shows notable compatibility information. It is not a comprehensive list of all compatible or incompatible products.
|DAQ Case 15
Current OS for the BaroVue10.
Description: Includes an SDI-12 example and a RS-232-serial program example. The programs are written for a CR1000X, but programs for other data loggers will be similar.
The SDI-12 example uses the C1! command to retrieve data. The program also adjusts the barometric pressure measurement to sea level. The adjustment is entered as a constant, Elevation_m, at the beginning of the program. Initially, the Elevation_m value is 1382; change this value to your site elevation, in meters.
The serial program demonstrates using RS-232 mode to initiate and return measurements using the SerialOut(), SerialIn(), and SplitStr() CRBasic instructions.
Number of FAQs related to BaroVue 10: 13
Expand AllCollapse All
It is possible that an older version of Short Cut is being used. Download the latest version of Short Cut.
The BaroVue 10 can fit inside an ENC100. For more information, refer to the Specifications information on the ENC100 product page.
There is no warm-up time for the BaroVue 10.
Contact your local Campbell Scientific office to purchase a calibrated BaroVue 10 sensor card (BVC10). After you receive the calibrated card, swap it with your current card, and mail it back to your local Campbell Scientific office. In this way, only one trip to the field site is required, and there will be no downtime for the sensor.
The BaroVue 10 has been designed to allow for zero downtime recalibration during normal use. Other sensors on the market need to be taken offline and sent back to the manufacturer to be recalibrated, often for weeks at a time, resulting in loss of data and costing time and money.
In comparison, the sensor card of the BaroVue 10 is calibrated ahead of time and sent out to customers to replace the old sensor card, requiring only one trip to the field and no downtime.
Further to this, the design of the BaroVue 10 allows customers to remove and replace their sensor card quickly and easily without having to disassemble the equipment in the enclosure or having to send the product back to the manufacturer. There is no need to turn the device off to replace the card.
Highly reliable measurements
The BaroVue 10 sensor card employs multiple sensor cells with statistical analysis to produce an accurate, precise, and repeatable measurement as well as a data-quality metric. The customer can use this quality metric for tracking calibration cycles, planning, maintenance schedules, and detecting pre-fault conditions.
The BaroVue 10 is a low power device and can be kept running continuously. The BaroVue 10 does not require any power cycling, unlike other products in this category.
Very low noise
The BaroVue 10 uses direct-to-digital sensors, and no analog-to-digital reconversion takes place inside the barometer.
These factors, coupled with the multiple sensor cells, result in a reduction in noise when compared with analog sensors.
Campbell Scientific aesthetics and form
The BaroVue 10 has been designed with functionality in mind, and its aesthetic complements other Campbell Scientific products.
The BaroVue 10 can be mounted in the orientation that best suits the application and has a smaller backplane footprint than its competitors.
While the measurement is equivalent, the BaroVue 10 is not a drop-in replacement. Unlike the CS100 and the CS106, the BaroVue 10 is a digital sensor that communicates either via SDI-12 or RS-232. To use the BaroVue 10 in a system that previously used a CS100/CS106, the wiring and program will need to be changed, as it is a digital sensor.
Campbell Scientific recommends calibration on a yearly basis. However, conformance to published standards or regulations may require this on a more frequent basis.
When using SDI-12 or RS-232 communication protocols, the pressure measurement can be returned in hPa, kPa, mmHg, inHg, or PSI. Refer to the BaroVue 10 manual for more information. If different measurement units are required, refer to a conversion chart.
Yes, but only if certain conditions are met:
In most cases, the enclosures or buildings where the BaroVue 10 is housed have enough ventilation for the pressure to stay at equilibrium with the ambient pressure. Therefore, the sensor does not need to be vented to the outside
However, if the sensor resides in an air-tight environment, and the pressure external to this environment needs to be measured, then the sensor will need to be vented to the outside. The preferred method for measuring the pressure external to the enclosure is to use a two-plate pressure port such as the RM Young 61002 to minimize dynamic pressure errors due to wind.