The ENC24/30 is a painted steel enclosure with additional wiring room for various components, including a datalogger, power supply, larger batteries, and communication and measurement peripherals. This enclosure, with a backplate with holes for mounting devices, can be ordered with one to four conduit openings or 12 cable entry seals. The ENC24/30 is designed for mounting on a tower, building, or other structure, but a user-supplied mount is required. This enclosure is available in two finishes. ENC24/30 is the mild-steel version, and ENC24/30S is the stainless-steel version.Read More
The ENC24/30's backplate is prepunched with half-inch-on-center holes suitable for attaching the datalogger, power supply, communication device, and measurement and control peripheral.
This enclosure is shipped with the 7363 enclosure supply kit that consists of desiccant, a humidity indicator card, cable ties, wire tie tabs, putty, grommets, screws, and PVC coupling. Additionally, Campbell Scientific offers a CS210 Enclosure Humidity Sensor for monitoring relative humidity inside of the enclosure. (See Ordering Info on the web page.)
|Construction||Painted, 14-gauge, mild steel with door gasket and stainless-steel hinges|
|Enclosure Classification||NEMA Type 3R, 4, and 12 (before being modified for cable entry)|
|Dimensions||61 x 76 x 20 cm (24 x 30 x 8 in.)|
|Weight||21 kg (46 lb)|
Note: The following shows notable compatibility information. It is not a comprehensive list of all compatible or incompatible products.
Number of FAQs related to ENC24/30: 2
There is not an easy way to calculate how much desiccant is needed in an enclosure or how quickly the desiccant will need to be replaced. The following conditions affect the replacement interval:
It may be advantageous to use more than one desiccant packet at a time inside an enclosure. Campbell Scientific recommends keeping an extra supply of desiccant on hand for as-needed replacement in the enclosure.
The desiccant should be replaced when the top colored dot on the humidity indicator card (pn 28878) turns pink; the humidity indicator card is included in the enclosure supply kit that is shipped with each enclosure. If the CS210, Enclosure Humidity Sensor, is used, replace the desiccant when the measured humidity exceeds 35%.