This enclosure is similar to the ENC10/12, except it has a raised lid. The raised lid allows taller peripherals to reside in it. The ENC10/12R typically houses a smaller datalogger, power supply, and a small peripheral. It can also house remotely located peripherals such as multiplexers.Read More
The ENC10/12R's backplate is prepunched with half-inch-on-center holes suitable for attaching a datalogger, power supply, and a communications or measurement and control peripheral.
The ENC10/12R is shipped with the 7363 enclosure supply kit that consists of desiccant, a humidity indicator card, cable ties, wire tie tabs, putty, grommets, screws, and PVC coupling. Additionally, Campbell Scientific offers a CS210 Enclosure Humidity Sensor for monitoring relative humidity inside of the enclosure. (See Ordering Info on the web page.)
|Construction||Fiberglass-reinforced polyester enclosure with door gasket, external grounding lug, stainless-steel hinge, and lockable hasps|
|Enclosure Classification||NEMA 4X (before being modified for cable entry, conduit, or ground lug)|
|Weight||4.1 kg (9 lb)|
Note: The following shows notable compatibility information. It is not a comprehensive list of all compatible or incompatible products.
|CR1000||Although the CR1000 is compatible, this datalogger can only fit inside the ENC10/12R if it is mounted vertically rather than horizontally.|
|CR3000||The ENC10/12R is compatible with the CR3000. Similar to the CR800/CR850 and CR1000, the CR3000 must be mounted vertically rather than horizontally. No space remains in the enclosure for additional peripherals.|
|CR800||Although the CR800 is compatible, this datalogger can only fit inside the ENC10/12R if it is mounted vertically rather than horizontally.|
|CR850||Although the CR850 is compatible, this datalogger can only fit inside the ENC10/12R if it is mounted vertically rather than horizontally.|
Retired peripherals that won’t fit in the same ENC10/12R enclosure as the datalogger/power supply include the AVW4, AM416, COM100, CR10KD, DC1765, and RC35.
Number of FAQs related to ENC10/12R: 14
Expand AllCollapse All
If you purchased a Stahlin enclosure from Campbell Scientific and it has a failed gasket, we can help you. We will need you to provide an image of the failed gasket, verify the size of your enclousre, and provide your contact information. We can then arrange to have a replacement gasket sent to you from Stahlin. For assistance, contact Technical Support at 435.227.9100.
There is not an easy way to calculate how much desiccant is needed in an enclosure or how quickly the desiccant will need to be replaced. The following conditions affect the replacement interval:
It may be advantageous to use more than one desiccant packet at a time inside an enclosure. Campbell Scientific recommends keeping an extra supply of desiccant on hand for as-needed replacement in the enclosure.
Backplates are available in sizes designed to fit in Campbell Scientific enclosures: 10 x 12 in., 12 x 14 in., 14 x 16 in., and 16 x 18 in.