The ENC24/30UL is a painted steel enclosure with additional wiring room for various components, including a data logger, power supply, larger batteries, and communication and measurement peripherals. It is UL508A listed, with a UL listed NEMA 4 rated enclosure and components. This enclosure, with a backplate with holes for mounting devices, can be ordered with one to four conduit openings or 12 cable entry seals. The ENC24/30UL is designed for mounting on a tower, building, or other structure, but a user-supplied mount is required. This enclosure is available in two finishes. ENC24/30UL is the mild-steel version, and ENC24/30SUL is the stainless-steel version.
To maintain its NEMA 4 rating, the ENC24/30UL does not come with a standard Campbell Scientific ground lug. Users must bring a ground wire inside the enclosure through a conduit or cable entry seal and then ground it to a ground bar, DIN-rail mounting bracket, or something similar.Read More
The backplate of the ENC24/30UL is prepunched with half-inch-on-center holes suitable for attaching the data logger, power supply, communication device, and measurement and control peripheral.
This enclosure is shipped with the 7363 enclosure supply kit that consists of desiccant, a humidity indicator card, cable ties, wire tie tabs, putty, grommets, screws, and PVC coupling. Additionally, Campbell Scientific offers a CS210 Enclosure Humidity Sensor for monitoring relative humidity inside of the enclosure. (See Ordering Info on the web page.)
|Construction||Painted, 14-gauge, mild steel with door gasket and stainless-steel hinges|
|Enclosure Classification||NEMA Type 3R, 4, and 12|
|Dimensions||61 x 76 x 20 cm (24 x 30 x 8 in.)|
|Weight||21 kg (46 lb)|
Note: The following shows notable compatibility information. It is not a comprehensive list of all compatible or incompatible products.
Number of FAQs related to ENC24/30UL: 2
There is not an easy way to calculate how much desiccant is needed in an enclosure or how quickly the desiccant will need to be replaced. The following conditions affect the replacement interval:
It may be advantageous to use more than one desiccant packet at a time inside an enclosure. Campbell Scientific recommends keeping an extra supply of desiccant on hand for as-needed replacement in the enclosure.
The desiccant should be replaced when the top colored dot on the humidity indicator card (pn 28878) turns pink; the humidity indicator card is included in the enclosure supply kit that is shipped with each enclosure. If the CS210, Enclosure Humidity Sensor, is used, replace the desiccant when the measured humidity exceeds 35%.