The AVW206 is a vibrating-wire analyzer module that includes an internal 910 to 918 MHz spread-spectrum radio for wireless communication, typically used in the U.S. and Canada. With this vibrating-wire analyzer module, your datalogger can measure vibrating-wire strain gages, pressure transducers, piezometers, tiltmeters, crackmeters, and load cells. These sensors are used in a wide variety of structural, hydrological, and geotechnical applications because of their stability, accuracy, and durability.Read More
The AVW206 uses vibrating-wire spectral-analysis technology (VSPECT™). VSPECT observes the incoming sensor signal, performs a Fourier transform and a spectral analysis (transforming the time series into indivdual sinusoidal components in the frequency spectrum), and determines the sensor frequency by identifying the largest signal in the acceptable range while filtering out environmental and electrical noise.
The AVW206 analyzer module also provides many self-checking diagnostics such as vibrating-element signal strength, signal-to-noise ratio, vibrating-element signal decay ratio, and incorrect signal response. These diagnostics can be running in the background to give continual feedback of the condition for each sensor.
The AVW206 typically transmits its data to an RF401 spread-spectrum radio that is connected to the datalogger. The AVW206 can also be connected directly to the datalogger.
|Internal Radio Frequency Range||910 to 918 MHz|
Older AVW206 modules (serial # < 11224) have 100 mW radios.
|Number of Vibrating-Wire Sensors Measured||Up to 2 vibrating-wire sensors can be connected to the analyzer module. Additional sensors can be measured by using an AM16/32-series multiplexer.|
|Power Requirements||9.6 to 16 Vdc|
|Analog Input/Outputs||2 differential (DF) vibrating-wire measurements (V+ and V-) and 2 single-ended (SE) ratiometric resistive half-bridge measurements (T+ and T-) for vibrating-wire sensor's onboard temperature sensor.|
|Digital Control Ports||
3 digital control ports (C1 – C3)
|RS-232 Port||1 9-pin RS-232 port (for connecting to a datalogger COM port)|
|Measurement Resolution||0.001 Hz RMS (±250 mV differential input range; -55° to +85°C)|
|Measurement Accuracy||±0.013% of reading (±250 mV differential input range; -55° to +85°C)|
|Input Voltage Range||±250 mV (differential)|
|Common Mode Range||±25 V|
|Baud Rates||Selectable from 1200 to 38.4 kbps (ASCII protocol is one start bit, one stop bit, eight data bits, and no parity.)|
|CE Compliance Standards to which Conformity Is Declared||IEC61326:2002|
|Dimensions||21.6 x 11.18 x 3.18 cm (8.5 x 4.5 x 1.2 in.)|
|Weight||0.43 kg (0.95 lb)|
Typical Current Drain @ 12 Vdc
|Quiescent, Radio Off||~0.3 mA|
|Radio Duty Cycling 1 s||~3 mA (includes quiescent current)|
|Radio Always On||~26 mA|
|Active RS-232 Communication||~6 mA|
The AVW206 Vibrating-Wire Spectrum Analyzer Module allows our dataloggers to measure vibrating-wire strain gages, pressure transducers, piezometers, tiltmeters, crackmeters, and load cells.
The AVW206's internal spread spectrum radio can transmit data to an RF401A, RF401, or RF430. The AVW206 does not have a transparent mode, and therefore its internal radio cannot communicate with an RF400 radio.
|CR200(X) Series||CR800/CR850||CR1000||CR3000||CR9000X||CR300 Series|
*The CR5000, CR10(X), and CR23X support the SDI-12 mode only.
The AVW206 requires a desiccated, non-condensing environment. A Campbell Scientific enclosure is recommended when the analyzer module is in the field. The AVW206 has built-in keyhole flanges for mounting to an enclosure backplate.
#If using the RS-232 port, the AVW206 must be mounted flat to fit in an ENC10/12, ENC10/12R, or ENC12/14.
The AVW206 is typically powered by the datalogger's power supply.
Newer modules with 250 mW radios must use OS 5 or higher for their operating system.
Number of FAQs related to AVW206: 14
Expand AllCollapse All
No. An AVW206, AVW211, or AVW216 is not capable of communicating with other AVW200-series modules.
Here is an example of the code that can be added to a datalogger program to acquire both the voltage of the AVW206 and the radio signal strength:
The recommended maximum is 24 AVW206 devices.
The AVW200-series modules are not required, but they do reduce the chance of erroneous readings by filtering electrical noise that is sometimes present.
The AVW200 must be wired directly to the datalogger via a cable.
The AVW206, AVW2011, and AVW216 have built-in spread-spectrum radios tuned to different frequencies that allow the datalogger to communicate with a remote AVW device through an appropriate radio link.
Two probable causes are the following:
It takes 2 s for the AVW200-series modules to acquire a measurement from each sensor. An AVW200-series module servicing two AM16/32B multiplexers, each with 16 vibrating-wire sensors, requires approximately 64 s to acquire data.
Use pn 17855, Data Cable, RS-232 Male to Pigtail.