With a CWS900E, you can use a variety of sensors in a wireless sensor network. (See the Compatibility information on the web page.) It contains an internal 868 MHz spread-spectrum radio that transmits data to the CWB100E Wireless Base Station or to another wireless sensor. The frequency of the CWS900E's internal radio is commonly used in Europe.Read More
The CWS900E has a sealed connector for attaching one rain gage, wind sensor, temperature/RH probe, or another low power sensor. It measures analog voltages, low level AC, pulse counts, and can supply an excitation voltage for powering some sensors or making bridge measurements. The type of measurement that the sensor performs is configured using Wireless Sensor Planner, Network Planner, or DevConfig software. The CWS900E interfaces with the PC for configuration via the A205 CWS Sensor to PC Interface.
The CWS900E can route its transmissions through up to three other wireless sensors. A datalogger is connected to the CWB100E base station for processing and storing the CWS900E's data.
The CWS900E is battery powered using either alkaline batteries or a rechargeable battery and a solar cell.
There are situations when it is desirable to make measurements in locations where the use of cabled sensors is problematic. Protecting cables by running them through conduit or burying them in trenches is time consuming, labor intensive, and sometimes not possible. Local fire codes may preclude the use of certain types of sensor cabling inside of buildings. In some applications measurements need to be made at distances where long cables decrease the quality of the measurement or are too expensive. There are also times when it is important to increase the number of measurements being made but the datalogger does not have enough available channels left for attaching additional sensor cables.
|Note: The internal radio is not designed to move a lot of data quickly. It takes 15 to 30 seconds per hop when moving data from a sensor, through a sensor used as a repeater, and ending up at the base radio. Going through three repeaters could take a data packet anywhere from 45 to 90 seconds to get to the base radio.|
|Weather Resistance||IP67 rating for sensor and battery pack (Battery pack must be properly installed. Each sensor is leak tested.)|
|Operating Temperature Range||-25° to +50°C|
|Operating Relative Humidity Range||0 to 100% (non-condensing)|
|Power Source||2 AA batteries with a battery life of 1 year assuming sensor samples taken every 10 minutes. (Optional solar charging available.)|
|Average Current Drain||300 μA (with 15-minute polling and depending on attached sensor)|
|Analog Input Range||-1 to +2.5 Vdc|
|Excitation Voltage||2.5 V, 3.3 V, 5.0 V (20 mA maximum)|
|Excitation Voltage Accuracy||±2% (-35° to +70°C)|
|Low-Level AC Input||20 mV minimum (10 kHz maximum frequency)|
|Analog Measurement Accuracy||
|Bridge Measurement Accuracy (+2.5 V excitation only)||
|Dimensions||15 x 6 x 4.5 cm (5.9 x 2.4 x 1.77 in.)|
|Weight||184 g (6.5 oz)|
|Maximum Count Rate||100 Hz|
|Minimum Open Time||5 ms|
|Minimum Closed Time||5 ms|
|Maximum Bounce Time||4 ms|
|Maximum Number of Counts||Records 1,000,000 counts before it rolls over.|
Internal 25 mW FHSS Radio
|Transmitter Power Output||25 mW (+14 dBm)|
|Receiver Sensitivity||-110 dBm (0.1% frame error rate)|
|Standby Typical Current Drain||3 μA|
|Receive Typical Current Drain||18 mA (full run)|
|Transmit Typical Current Drain||45 mA|
|Average Operating Current||15 μA (with 1-second access time)|
|Quality of Service Management||RSSI|
|Additional Features||GFSK modulation, data interleaving, forward error correction, data scrambling, RSSI reporting|
None of our retired dataloggers are compatible.
One sensor can be connected to the CWS900E. Sensors with the -CWS option will include a connector that mates with the CWS900E. Alternatively, a sensor can connect with the CWS900E via the A150 Desiccant Case (see bottom of page).
The A150 Desiccated Case has a -CWS cable termination option. It allows our CS450 and CS455 pressure transducers or other sensors that have a vented cable to be connected to the CWS900E. Sensors that measure analog voltages, frequency, and pulse can also use the A150 to connect with the CWS900E.
Latest firmware for the CWS900.
The Wireless Sensor Planner is a tool for use with Campbell Scientific wireless sensors. It assists in designing and configuring wireless sensor networks.
Number of FAQs related to CWS900E: 4
Expand AllCollapse All
Yes. For more information, refer to the Wireless Sensor Network Instruction Manual.
CWS900-series interfaces are not designed for use inside enclosures.
To incorporate a sensor that is compatible with wireless sensor interfaces into a wireless network, a CWS900-series wireless sensor interface is needed, as well as an A205 CWS-to-PC interface to configure it.