The 034B, manufactured by Met One, combines a 3-cup anemometer and vane into a single integrated package to measure wind speed and direction. It is cabled for use with our dataloggers, and can provide measurements for a variety of applications.Read More
The "-L" on a product model indicates that the cable length is specified at the time of order.
The 034B monitors wind speed using a three-cup anemometer that contains a sealed magnetic reed switch. Rotation of the cup wheel produces a pulse that is directly proportional to wind speed. The frequency of the pulse is measured by the datalogger pulse count channel, then converted to engineering units (mph, m/s, knots).
Wind direction is sensed with a potentiometer. With the precision excitation voltage from the datalogger applied to the potentiometer element, the output signal is an analog voltage that is directly proportional to the azimuth of the wind direction.
|Operating Temperature Range||-30° to +70°C|
|Weight||907 g (2.0 lb)|
Wind Speed (Anemometer)
|Range||0 to 75 m/s (0 to 167 mph)|
|Accuracy < 10.14 m/s (22.7 mph)||0.1 m/s (0.25 mph)|
|Accuracy > 10.14 m/s (22.7 mph)||±1.1% of true|
|Resolution||(1.789 mph) / (scan rate in seconds) or (0.7998 m s–1) / (scan rate in seconds)|
|Starting Threshold||0.4 m/s (0.9 mph)|
|Sensor Output||Pulsed contact closure|
|Anemometer Radius||10.7 cm (4.2 in.)|
|Anemometer Height||24.4 cm (9.6 in.)|
|Output Signal||Contact closure (reed switch)|
Wind Direction (Vane)
|Electrical Range||356° (4° open)|
|Potentiometer Resistance||0 to 10 kΩ (open at crossover)|
|Threshold||0.4 m s–1 (0.9 mph)|
|Vane Length||33.5 cm (13.2 in.)|
The 034B can be attached to a crossarm via a 17953 NU-RAIL fitting or a CM220 Right Angle Mounting Bracket. Alternatively, the 034B can be attached to the top of a CM106B or stainless-steel tripod via the CM216 Sensor Mounting Kit. The CM216 extends 4 in. above the mast of the tripod.
The 034B's anemometer uses one pulse count channel on the datalogger. Its wind vane requires one single-ended channel and access to an excitation channel (the excitation channel can be shared with other high impedance sensors).
The 034B's anemometer is measured by the PulseCount Instruction in CRBasic, and by Instruction 3 (Pulse Count) in Edlog. The wind vane is measured by the BrHalf Instruction in CRBasic and by Instruction 4 (Excite-Delay-SE) in Edlog. The measurements are typically processed for output with the Wind Vector instruction (not present in the CR500 or CR9000).
*Measurements are typically processed for output with the Wind Vector instruction, which is not present in the CR500 or CR9000.
Number of FAQs related to 034B-L: 15
Expand AllCollapse All
Wind vane tails are user-replaceable and are available for purchase as pn 13765, Replacement Wind Vane Tail for 034B or 034A.
No. A datalogger is needed to measure and store data from the 034B-L.
For proper siting of the wind sensor, refer to the 034B-L manual. The 034B-L can be mounted using a 1.00 IPS pipe of any length.
The supply voltage is 5 Vdc.
They are completely interchangeable, and no changes are needed.
The sensor does not consume power to operate.
The 034B-L uses a switch closure for wind speed measurement, and the 03002-L uses a low-level ac output for wind speed measurement. Both use a pulse channel on the datalogger to measure wind speed.
The information included on a calibration sheet differs with each sensor. For some sensors, the sheet contains coefficients necessary to program a datalogger. For other sensors, the calibration sheet is a pass/fail report.
Most Campbell Scientific sensors are available as an –L, which indicates a user-specified cable length. If a sensor is listed as an –LX model (where “X” is some other character), that sensor’s cable has a user-specified length, but it terminates with a specific connector for a unique system:
If a sensor does not have an –L or other –LX designation after the main model number, the sensor has a set cable length. The cable length is listed at the end of the Description field in the product’s Ordering information. For example, the 034B-ET model has a description of “Met One Wind Set for ET Station, 67 inch Cable.” Products with a set cable length terminate, as a default, with pigtails.
If a cable terminates with a special connector for a unique system, the end of the model number designates which system. For example, the 034B-ET model designates the sensor as a 034B for an ET107 system.