The TE525WS, manufactured by Texas Electronics, is a tipping bucket rain gage that conforms to the National Weather Service recommendation for an 8 in. funnel orifice. It measures rainfall in 0.01 in. increments. This tipping bucket is compatible with all Campbell Scientific dataloggers, and it is widely used in environmental monitoring applications.Read More
The "-L" on a product model indicates that the cable length is specified at the time of order.
The TE525WS funnels precipitation into a bucket mechanism that tips when filled to its calibrated level. A magnet attached to the tipping mechanism actuates a switch as the bucket tips. The momentary switch closure is counted by the pulse-counting circuitry of our dataloggers.
|Sensor Type||Tipping bucket/magnetic reed switch|
|Operating Temperature Range||0° to 50°C|
|Volume per Tip||8.25 ml/tip (0.28 fl. oz/tip)|
|Rainfall per Tip||0.254 mm (0.01 in.)|
|Accuracy||1.0% up to 2 in./h (50 mm/h)|
|Cable Type||2-conductor shielded|
|Funnel Collector Diameter||20.3 cm (8 in.)|
|Height||26.7 cm (10.5 in.)|
|Cable Weight||0.1 kg (0.2 lb) per 3.05-m (10-ft) length|
|Tipping Bucket Weight||1.0 kg (2.2 lb)|
The TE525WS rain gage mounts to a CM300-series Mounting Pole or a user-supplied 1.5” IPS pole. Several pedestal options are available to secure a CM300-series pole to the ground (see Ordering Information). Accurate measurements require the gage to be level.
The TE525WS is directly compatible with the CS705 Snowfall Conversion Adapter. The CS705 uses PG:E antifreeze to melt snow, allowing the TE525WS to measure the water content of snow.
Campbell Scientific offers the 260-953 Wind Screen to help minimize the affect of wind on the rain measurements. This wind screen consists of 32 leaves that hang freely and swing as the wind moves past them.
Number of FAQs related to TE525WS-L: 14
Expand AllCollapse All
The accuracy of the TE525WS-L diminishes as rainfall intensity increases (see the Accuracy specification) because at higher intensities, rain will keep pouring into the tipping mechanism as it is tipping, causing missed tips. The maximum listed intensity in the specifications is 2 to 3 in./hr, which would give an accuracy of +0%, -3.5%.
Replacement screens and funnels can be ordered by calling Campbell Scientific.
A TE525-L can be converted to a TE525MM-L and vice versa because the inner tipping mechanisms are the same. To convert a TE525-L or TE525MM-L to a TE525WS-L, the rain gage must be sent in to Campbell Scientific to change the tipping mechanism. If an 8 inch funnel is used on a TE525-L or TE525MM-L without changing the tipping mechanism, the multipliers for the rain measurement will change. For information about these multipliers, see the TE525 Instruction Manual.
Not enough tips or too many tips.
The 260-953 Alter-type rain gage wind screen consists of 32 heavy metal leaves that hang freely and swing as the wind moves past them. The swinging leaves act as a wind damper and help minimize the effect of wind on the rain measurements without adding additional turbulence.
When a rain gage is out of calibration, it is usually because of the buildup of dirt and grime on the internal surfaces of the tipping bucket mechanism. Cleaning the internal surfaces usually brings the rain gage back into calibration. It is also possible that a rain gage is out of calibration because it is no longer level.
To minimize the possible occurrence of calibration errors, perform routine cleaning and maintenance of the rain gage at least once every three months. The environmental conditions at a particular site may require a facility to perform cleaning, leveling, and maintenance on a much more frequent schedule.
The most common errors are either that the rain gage appears to have drifted out of calibration or that the tips are not being correctly recorded by the datalogger.
The information included on a calibration sheet differs with each sensor. For some sensors, the sheet contains coefficients necessary to program a datalogger. For other sensors, the calibration sheet is a pass/fail report.
Not every sensor has different cable termination options. The options available for a particular sensor can be checked by looking in two places in the Ordering information area of the sensor product page:
If a sensor is offered in an –ET, –ETM, –LC, –LQ, or –QD version, that option’s availability is reflected in the sensor model number. For example, the 034B is offered as the 034B-ET, 034B-ETM, 034B-LC, 034B-LQ, and 034B-QD.
All of the other cable termination options, if available, are listed on the Ordering information area of the sensor product page under “Cable Termination Options.” For example, the 034B-L Wind Set is offered with the –CWS, –PT, and –PW options, as shown in the Ordering information area of the 034B-L product page.
Note: As newer products are added to our inventory, typically, we will list multiple cable termination options under a single sensor model rather than creating multiple model numbers. For example, the HC2S3-L has a –C cable termination option for connecting it to a CS110 instead of offering an HC2S3-LC model.