The HMP155A provides reliable relative humidity (RH) and temperature measurements for a wide range of applications. It uses a HUMICAP®180R capacitive thin film polymer sensor to measure RH over the 0 to 100% RH range. A PRT measures temperature over the -80° to +60°C range. This rugged, accurate temperature/RH probe is manufactured by Vaisala.Read More
The "-L" on a product model indicates that the cable length is specified at the time of order.
To reduce the current drain, power can be supplied to the HMP155A only during measurement when the sensor is connected to the data logger’s switched 12 V terminal. Data loggers that do not have a switched 12 V terminal, such as the CR510 or CR7, can use the SW12V switched 12 V device to switch power to the sensor only during measurement.
|Electromagnetic Compatibility||Complies with EMC standard EN61326-1 Electromagnetic|
|Filter Description||Sintered PTFE|
|Voltage Output||0 to 1 Vdc|
|Average Current Consumption||≤ 3 mA (analog output mode)|
|Operating Voltage||7 to 28 Vdc|
|Settling Time||2 s (at power up)|
|Tip Diameter||1.2 cm (0.5 in.)|
|Length||27.9 cm (11 in.)|
|Head Height||4 cm (1.6 in.)|
|Body Height||2.4 cm (0.9 in.)|
|Body Width||2.0 cm (0.8 in.)|
|Measurement Range||0.8 to 100% RH (non-condensing)|
|Factory Calibration Uncertainty||
|-NOTE-||Accuracy specifications include non-linearity, hysteresis, and repeatability.|
|Accuracy at 15° to 25°C||
|Accuracy at -60° to -40°C||±(1.4 + 0.032 × reading) % RH|
|Accuracy at -40° to -20°C||±(1.2 + 0.012 × reading) % RH|
|Accuracy at -20° to +40°C||±(1.0 + 0.008 × reading) % RH|
|Accuracy at 40° to 60°C||± (1.2 + 0.012 × reading) % RH|
|Sensor||Pt 100 RTD 1/3 class B IEC 751|
|Measurement Range||-80° to +60°C|
|Accuracy with Voltage Output||
|Entire Temperature Range||Refer to graph in probe manual.|
Note: The following shows notable compatibility information. It is not a comprehensive list of all compatible or incompatible products.
When exposed to sunlight, the HMP155A must be housed in a 41005-5 or RAD14 14-plate naturally aspirated radiation shield. The 41005-5 and RAD14 attach to a crossarm, mast, or user-supplied pipe with a 2.5 to 5.3 cm (1.0 to 2.1 in.) outer diameter. The RAD14 uses a double-louvered design that offers improved sensor protection from driving rain, snow, and insect intrusion, It also has lower self-heating in bright sunlight combined with higher temperatures (> 24°C [~75°F])
and low wind speeds (< 2 m s-1 [~4.5 mph]), giving a better measurement.
Number of FAQs related to HMP155A-L: 7
Expand AllCollapse All
The maximum cable length is 304 m (1000 ft).
No. The HMP155A-L requires the 41005-5 14-Plate Solar Radiation Shield.
Most Campbell Scientific sensors are available as an –L, which indicates a user-specified cable length. If a sensor is listed as an –LX model (where “X” is some other character), that sensor’s cable has a user-specified length, but it terminates with a specific connector for a unique system:
If a sensor does not have an –L or other –LX designation after the main model number, the sensor has a set cable length. The cable length is listed at the end of the Description field in the product’s Ordering information. For example, the 034B-ET model has a description of “Met One Wind Set for ET Station, 67 inch Cable.” Products with a set cable length terminate, as a default, with pigtails.
If a cable terminates with a special connector for a unique system, the end of the model number designates which system. For example, the 034B-ET model designates the sensor as a 034B for an ET107 system.
Not every sensor has different cable termination options. The options available for a particular sensor can be checked by looking in two places in the Ordering information area of the sensor product page:
If a sensor is offered in an –ET, –ETM, –LC, –LQ, or –QD version, that option’s availability is reflected in the sensor model number. For example, the 034B is offered as the 034B-ET, 034B-ETM, 034B-LC, 034B-LQ, and 034B-QD.
All of the other cable termination options, if available, are listed on the Ordering information area of the sensor product page under “Cable Termination Options.” For example, the 034B-L Wind Set is offered with the –CWS, –PT, and –PW options, as shown in the Ordering information area of the 034B-L product page.
Note: As newer products are added to our inventory, typically, we will list multiple cable termination options under a single sensor model rather than creating multiple model numbers. For example, the HC2S3-L has a –C cable termination option for connecting it to a CS110 instead of offering an HC2S3-LC model.
Note the difference between calibration and a field check. Calibration cannot be done in the field, as it requires an experienced technician and specialized equipment.
Field checks of measurements can be done to determine if the data make sense with the real-world conditions. Follow these steps to field check a sensor:
Many Campbell Scientific sensors are available with different cable termination options. These options include the following:
Note: The availability of cable termination options varies by sensor. For example, sensors may have none, two, or several options to choose from. If a desired option is not listed for a specific sensor, contact Campbell Scientific for assistance.
Single-ended channels are typically used when it is desirable to use fewer channels and the cable length is shorter—less than 6.1 m (20 ft). Differential channels are preferable when the cable needs to be longer than 6.1 m (20 ft).