The CMP21, manufactured by Kipp & Zonen, is an ISO secondary-standard pyranometer with an internal thermistor. It monitors solar radiation for the full solar spectrum range, and is well-suited for scientific use and in top-level solar-radiation monitoring networks. This pyranometer connects directly to a Campbell Scientific datalogger.Read More
The CMP21 measures solar radiation with a high-quality blackened thermopile protected by two glass domes. An internal thermistor allows individually optimized temperature compensation of the measurements.
The CMP21 has flat spectral sensitivity, from 285 to 2800 nm, that makes it ideal for applications in natural sunlight, under plant canopies, in green houses or buildings, and inverted to measure reflected solar radiation.
A desiccant-filled drying cartridge prevents dew from forming on the inner sides of the CMP21's domes. Campbell Scientific also offers the CVF4 heater/ventilator that keeps its domes free from ice and dew.
The CMP21 includes a white snap-on sun shield that reduces the sensor's temperature. A bubble level and adjusting leveling screws enable the sensor to be leveled without using a leveling base.
|ISO Classification||Secondary Standard|
|Spectral Range||285 to 2800 nm|
|Sensitivity Range||7 to 14 µV/W/m2|
|Temperature Dependence of Sensitivity||< 1% (-20° to +50°C)|
|Response Time||< 5 s (95% of final value)|
|Zero Offset Due to Thermal Radiation||< 7 W/m2 (200 W/m2)|
|Non-Stability||< 0.5% (change/year)|
|Non-Linearity||< 0.2% (0 to 1000 W/m2)|
|Directional Error||< 10 W/m2 (up to 80° with 1000 W/m2 beam)|
|Tilt Error||< 0.2%|
|Impedance||10 to 100 Ω|
|Operating Temperature Range||-40° to +80°C|
|Typical Signal Output||0 to 15 mV (for atmospheric applications)|
|Maximum Irradiance||4000 W/m2|
|Expected Daily Uncertainty||< 2%|
|Dome Diameter||5 cm (2 in.)|
|Width||15 cm (5.9 in.) with shield|
|Height||9.25 cm (3.64 in.)|
|Weight||0.9 kg (2 lb) with 33 ft cable|
The CMP21 has a bubble level and two leveling feet, which allow it to be leveled without using a leveling base. This pyranometer mounts to a mast, crossarm, or pole (1.0 in. to 2.1 in. OD) via the CM255 or CM255LS mounting stand. The CMP21 should be mounted away from all obstructions and reflective surfaces that might adversely effect the measurement.
|CR200(X) Series||CR800/CR850||CR1000||CR3000||CR9000X||CR6 Series||CR300 Series|
Number of FAQs related to CMP21-L: 9
Expand AllCollapse All
A cable length of 300 ft can be used with the understanding that additional noise may be introduced into the measurement. Typically, in systems with pyranometers, the sensors are installed near the datalogger (within 50 ft). If an application requires long leads, contact Campbell Scientific for assistance.
No. These are all passive devices that create their own voltage signal as a response to the measurement being made. There is, however, a heater and ventilator for these sensors (CVF4-L) that requires a power supply.
Not every sensor has different cable termination options. The options available for a particular sensor can be checked by looking in two places in the Ordering information area of the sensor product page:
If a sensor is offered in an –ET, –ETM, –LC, –LQ, or –QD version, that option’s availability is reflected in the sensor model number. For example, the 034B is offered as the 034B-ET, 034B-ETM, 034B-LC, 034B-LQ, and 034B-QD.
All of the other cable termination options, if available, are listed on the Ordering information area of the sensor product page under “Cable Termination Options.” For example, the 034B-L Wind Set is offered with the –CWS, –PT, and –PW options, as shown in the Ordering information area of the 034B-L product page.
Note: As newer products are added to our inventory, typically, we will list multiple cable termination options under a single sensor model rather than creating multiple model numbers. For example, the HC2S3-L has a –C cable termination option for connecting it to a CS110 instead of offering an HC2S3-LC model.
Most Campbell Scientific sensors are available as an –L, which indicates a user-specified cable length. If a sensor is listed as an –LX model (where “X” is some other character), that sensor’s cable has a user-specified length, but it terminates with a specific connector for a unique system:
If a sensor does not have an –L or other –LX designation after the main model number, the sensor has a set cable length. The cable length is listed at the end of the Description field in the product’s Ordering information. For example, the 034B-ET model has a description of “Met One Wind Set for ET Station, 67 inch Cable.” Products with a set cable length terminate, as a default, with pigtails.
If a cable terminates with a special connector for a unique system, the end of the model number designates which system. For example, the 034B-ET model designates the sensor as a 034B for an ET107 system.
Yes. Each pyranometer is shipped with an instruction manual provided by Kipp & Zonen that contains information concerning its construction, spectral sensitivity, cosine response, and a simple sensor check out procedure. Included with the sensor and manual is a calibration certificate with the sensor sensitivity value and serial number.
Because of the loss of IR radiation, nearly all thermopile instruments typically have a negative offset. This offset is most easily visible at night-time, when a small negative value is read instead of zero. This same offset is present during the daytime, but it is not as visible because of the large solar signal.
Another common issue involves leveling an instrument. Leveling a thermopile instrument can cause errors in the direct beam component because the cosine response is not correct. These errors are more notable when the sun is close to the horizon because the angle is so shallow.
Many Campbell Scientific sensors are available with different cable termination options. These options include the following:
Note: The availability of cable termination options varies by sensor. For example, sensors may have none, two, or several options to choose from. If a desired option is not listed for a specific sensor, contact Campbell Scientific for assistance.
All of these pyranometers measure the amount of solar short-wave radiation incident on a surface. Because of their accuracy and performance level, the CMP11-L and CMP21-L are classified by the International Organization for Standardization (ISO) and World Meteorological Organization (WMO) as Secondary Standard sensors, representing the highest level of measurements made by pyranometers. In comparison, the CMP6-L is classified as a First Class sensor, which is the middle classification for pyranometers. (Second Class is the lowest pyranometer classification.)