The CS210 enclosure humidity sensor measures relative humidity levels inside environmental enclosures. The sensor is housed inside the enclosure. A Campbell Scientific data logger measures this sensor and stores its data.Read More
When CS210's measurements indicates that the RH level inside the enclosure is 40% or more, the desiccant should be replaced with another packet.
By having the data logger measure the sensor, the measured values can be incorporated into the transmitted data stream of remotely located data loggers. The probe has an 11-in. (28 cm) cable length, which should be adequate to place the probe near the enclosure conduits.
|Sensor||Elan HM2000 series precision bulk-polymer|
|Relative Humidity Measurement Range||0 to 100% (non-condensing)|
|RH Output Signal Range||0 to 1.0 Vdc|
|Accuracy||±3% RH (10 to 90% RH) (at 25°C)|
|Typical Long-Term Stability||Better than 3% RH per year|
|Response Time||10 seconds for a 30% to 90% RH step change (at 25°C, 90% response)|
|Current Consumption||< 0.5 mA|
|Supply Voltage||5 ± 0.25 Vdc|
|Settling Time||10 s|
|Operating Temperature Range||0° to 50°C|
|Cable Length||27.9 cm (11 in.)|
|Sensor Head Dimensions||2.5 x 1.3 x 0.8 cm (1.0 x 0.5 x 0.3 in.)|
|Weight||9 g (0.3 oz)|
Note: The following shows notable compatibility information. It is not a comprehensive list of all compatible or incompatible products.
Number of FAQs related to CS210: 6
Expand AllCollapse All
Yes. The sensor can be used with a multiplexer that is capable of making a single-ended voltage measurement.
The CS210 does not have any user-serviceable parts, and it does not require any routine maintenance.
Campbell Scientific recommends replacing the CS210 after every three to five years of continuous use. Alternatively, if the CS210 probe fails, replace the sensor at that time.
Incorrect wiring, program errors, or a failed sensor might be the cause.
The desiccant should be replaced when the top colored dot on the humidity indicator card (pn 28878) turns pink; the humidity indicator card is included in the enclosure supply kit that is shipped with each enclosure. If the CS210, Enclosure Humidity Sensor, is used, replace the desiccant when the measured humidity exceeds 35%.
Replace the desiccant packs when the enclosure’s RH level exceeds 40%.