The ENC16/18 is a weather-resistant enclosure that is 16 inches wide and 18 inches tall. It can house a datalogger, a power supply, and at least one peripheral. This enclosure is recommended for applications that need to have multiple communication or measurement and control peripherals housed in the same enclosure.Read More
The ENC16/18's backplate is prepunched with half-inch-on-center holes suitable for attaching a datalogger, power supply, and a communications or measurement and control peripheral.
This enclosure is shipped with the 7363 enclosure supply kit that consists of desiccant, a humidity indicator card, cable ties, wire tie tabs, putty, grommets, screws, and PVC coupling. Additionally, Campbell Scientific offers a CS210 Enclosure Humidity Sensor for monitoring relative humidity inside of the enclosure. (See Ordering Info on the web page.)
|Color||White (Reflects solar radiation, reducing temperature gradients inside the enclosure without using a separate radiation shield.)|
|Construction||Fiberglass-reinforced polyester enclosure with door gasket, external grounding lug, stainless-steel hinge, and lockable hasps|
|Enclosure Classification||NEMA 4X (before being modified for cable entry)|
|Weight||7.7 kg (17 lb)|
Note: The following shows notable compatibility information. It is not a comprehensive list of all compatible or incompatible products.
The -VC cable opening option is required to fit a BP84 84-Ahr power supply in the ENC16/18 (see Ordering Info).
Number of FAQs related to ENC16/18: 18
Expand AllCollapse All
If you purchased a Stahlin enclosure from Campbell Scientific and it has a failed gasket, we can help you. We will need you to provide an image of the failed gasket, verify the size of your enclousre, and provide your contact information. We can then arrange to have a replacement gasket sent to you from Stahlin. For assistance, contact Technical Support at 435.227.9100.
Most likely, yes. The datalogger is protected against electrical noise, and 60 Hz filtering can be selected in many sensor measurements to filter out the effects of 60 Hz noise.
There is not an easy way to calculate how much desiccant is needed in an enclosure or how quickly the desiccant will need to be replaced. The following conditions affect the replacement interval:
It may be advantageous to use more than one desiccant packet at a time inside an enclosure. Campbell Scientific recommends keeping an extra supply of desiccant on hand for as-needed replacement in the enclosure.