The TE525WS-LC conforms to the National Weather Service recommendation for an 8" funnel orifice. It measures rainfall in 0.01 inch increments. The TE525WS-LC funnels precipitation into a bucket mechanism that tips when filled to a calibrated level.Read More
The "-LC" version of this product terminates in a connector (instead of pigtails) for use with a CS110 Electric Field Meter or MetData1 Weather Station. The length of the cable is specified at the time of order.
The cable on this version of the TE525WS tipping bucket rain gage has a connector rather than the standard pigtails. The connector attaches to the connector panel on a CS110 Electric Field Meter or MetData1 Weather Station.
The TE525WS-LC conforms to the National Weather Service recommendation for an 8" funnel orifice. It measures rainfall in 0.01 inch increments. The TE525WS-LC funnels precipitation into a bucket mechanism that tips when filled to a calibrated level.
|Sensor Type||Tipping bucket/magnetic reed switch|
|Temperature||0° to 50°C|
|Volume per Tip||8.25 ml/tip (0.28 fl. oz/tip)|
|Rainfall per Tip||0.254 mm (0.01 in.)|
|Accuracy||1.0% up to 2 in./h (50 mm/h)|
|Funnel Collector Diameter||20.3 cm (8 in.)|
|Cable||2-conductor shielded cable|
|Height||26.7 cm (10.5 in.)|
|Tipping Bucket Weight||1.0 kg (2.2 lb)|
|Cable Weight||0.1 kg (0.2 lb) per 3.048-m (10-ft) length|
The TE525WS is compatible with our CS705, enabling measurement of snowfall.
The connector can be attached to the CS110 Electric Field Meter or MetData1 Weather Station. The CS110 has a CR1000 and the MetData1 has an on-board CR10X.
Number of FAQs related to TE525WS-LC: 5
Expand AllCollapse All
The information included on a calibration sheet differs with each sensor. For some sensors, the sheet contains coefficients necessary to program a datalogger. For other sensors, the calibration sheet is a pass/fail report.
Not every sensor has different cable termination options. The options available for a particular sensor can be checked by looking in two places in the Ordering information area of the sensor product page:
If a sensor is offered in an –ET, –ETM, –LC, –LQ, or –QD version, that option’s availability is reflected in the sensor model number. For example, the 034B is offered as the 034B-ET, 034B-ETM, 034B-LC, 034B-LQ, and 034B-QD.
All of the other cable termination options, if available, are listed on the Ordering information area of the sensor product page under “Cable Termination Options.” For example, the 034B-L Wind Set is offered with the –CWS, –PT, and –PW options, as shown in the Ordering information area of the 034B-L product page.
Note: As newer products are added to our inventory, typically, we will list multiple cable termination options under a single sensor model rather than creating multiple model numbers. For example, the HC2S3-L has a –C cable termination option for connecting it to a CS110 instead of offering an HC2S3-LC model.
Most Campbell Scientific sensors are available as an –L, which indicates a user-specified cable length. If a sensor is listed as an –LX model (where “X” is some other character), that sensor’s cable has a user-specified length, but it terminates with a specific connector for a unique system:
If a sensor does not have an –L or other –LX designation after the main model number, the sensor has a set cable length. The cable length is listed at the end of the Description field in the product’s Ordering information. For example, the 034B-ET model has a description of “Met One Wind Set for ET Station, 67 inch Cable.” Products with a set cable length terminate, as a default, with pigtails.
If a cable terminates with a special connector for a unique system, the end of the model number designates which system. For example, the 034B-ET model designates the sensor as a 034B for an ET107 system.
This depends on the information contained in the calibration sheet:
Many Campbell Scientific sensors are available with different cable termination options. These options include the following:
Note: The availability of cable termination options varies by sensor. For example, sensors may have none, two, or several options to choose from. If a desired option is not listed for a specific sensor, contact Campbell Scientific for assistance.