The SDM-CD16D increases the number of digital outputs that can be controlled by a Campbell Scientific datalogger. The SDM-CD16D is commonly used to drive normal logic level inputs.Read More
The SDM-CD16D increases the number of digital outputs that can be controlled (i.e., set to 0 or 5 V) by a Campbell Scientific datalogger. In addition to being able to drive normal logic level inputs, when an output is set HI, a boost circuit allows it to source a current of up to 100 mA for controlling low voltage valves, relays, or other devices.
Up to 15 SDM-CD16Ds may be addressed, making it possible to control a maximum of 240 ports from the first three datalogger control ports.
The SDM Jumper Wire Kit (pn 32505) connects up to four SDMs to the datalogger. This kit is recommended when multiple SDMs are connected to one datalogger or for extremely short distances between the SDM and datalogger. The CABLE5CBL-L cable is recommended for connecting a single SDM to the datalogger, and for longer distances between the SDM and datalogger.
|Operating Temperature||-25° to +70°C|
|Operating Voltage||9 to 18 Vdc|
|Current Drain||100 µA typical (all ports HI, no load)|
|EMC Status||Complies with EN55022-1:1998. and EN50082-1:1998.|
|Output Voltage (no load)||Output ON/HI, nominal 5 V (minimum 4.5 V)|
|Output OFF/LO||Nominal 0 V (maximum 0.1 V)|
|Output Source Current||Output sources 36 mA @ 3 V, 115 mA short-circuited to ground|
|Maximum Output Current (total all outputs)||
400 mA and 50°C and 12 V supply
(The maximum current [total all outputs] should be derated by 50 mA for every 10°C above 50°C and/or 50 mA for every voltage above 12 V.)
|Weight||318 g (11 oz)|
Up to 15 SDM devices can be connected to a single datalogger. To communicate with connected SDM devices, the CR3000 and CR5000 use the SDM-C1, SDM-C2, and SDM-C3 ports. The other dataloggers use ports labeled C1, C2, and C3. The 21X also requires channel 1H.
The SDMCD16AC Instruction controls SDM-CD16D operation in CRBasic. Edlog Instruction 29 controls SDM-CD16D operation in the CR7, and Edlog Instruction 104 controls SDM-CD16D operation in the CR10(X), CR23X, and 21X dataloggers.
*For the CR3000 and CR5000, SDMs are connected to the ports labeled SDM-C1, SDM-C2, and SDM-C3.
**On the CR7X, SDMs connect to the C1, C2, and C3 terminals provided on the CR700X control module (after Sept 1996).
The maximum output current that the SDM-CD16D can source is 400 mA. The power requirements are large compared to most Campbell Scientific products when driving significant loads. For many applications, an external rechargeable power supply is recommended to power the SDM-CD16D. This power supply should share a common grounding point with the datalogger. For some applications, the datalogger's sealed rechargeable battery may suffice; in this case, float-charging the batteries from an ac source is recommended.
The SDM-CD16D requires a desiccated, non-condensing environment; a Campbell Scientific enclosure is recommended. Grommets and screws are provided to attach the flanges/bracket to the backplate of our enclosures.