The CS655 is a multiparameter smart sensor that uses innovative techniques to monitor soil volumetric-water content, bulk electrical conductivity, and temperature. It outputs an SDI-12 signal that many of our data loggers can measure. It has shorter rods than the CS650, for use in problem soils.
Note: The cable termination options for this sensor are not suitable for use with an ET107 station. For this type of station, use the CS655-LC sensor instead, which has a suitable cable connector.Read More
The CS655 consists of two 12-cm-long stainless steel rods connected to a printed circuit board. The circuit board is encapsulated in epoxy and a shielded cable is attached to the circuit board for data logger connection.
The CS655 measures propagation time, signal attenuation, and temperature. Dielectric permittivity, volumetric water content, and bulk electrical conductivity are then derived from these raw values.
Measured signal attenuation is used to correct for the loss effect on reflection detection and thus propagation time measurement. This loss-effect correction allows accurate water content measurements in soils with bulk EC ≤8 dS m-1 without performing a soil-specific calibration.
Soil bulk electrical conductivity is also calculated from the attenuation measurement. A thermistor in thermal contact with a probe rod near the epoxy surface measures temperature. Horizontal installation of the sensor provides accurate soil temperature measurement at the same depth as the water content. Temperature measurement in other orientations will be that of the region near the rod entrance into the epoxy body.
|Measurements Made||Soil electrical conductivity (EC), relative dielectric permittivity, volumetric water content (VWC), soil temperature|
|Required Equipment||Measurement system|
|Soil Suitability||Short rods are easy to install in hard soil. Suitable for soils with higher electrical conductivity.|
|Sensing Volume||3600 cm3 (~7.5 cm radius around each probe rod and 4.5 cm beyond the end of the rods)|
|Electromagnetic||CE compliant (Meets EN61326 requirements for protection against electrostatic discharge and surge.)|
|Operating Temperature Range||-50° to +70°C|
|Sensor Output||SDI-12; serial RS-232|
|Warm-up Time||3 s|
|Measurement Time||3 ms to measure; 600 ms to complete SDI-12 command|
|Power Supply Requirements||6 to 18 Vdc (Must be able to supply 45 mA @ 12 Vdc.)|
|Maximum Cable Length||610 m (2000 ft) combined length for up to 25 sensors connected to the same data logger control port|
|Rod Spacing||32 mm (1.3 in.)|
|Ingress Protection Rating||IP68|
|Rod Diameter||3.2 mm (0.13 in.)|
|Rod Length||120 mm (4.7 in.)|
|Probe Head Dimensions||85 x 63 x 18 mm (3.3 x 2.5 x 0.7 in.)|
|Cable Weight||35 g per m (0.38 oz per ft)|
|Probe Weight||240 g (8.5 oz) without cable|
|Active (3 ms)||
|Quiescent||135 µA typical (@ 12 Vdc)|
|Range for Solution EC||0 to 8 dS/m|
|Range for Bulk EC||0 to 8 dS/m|
|Accuracy||±(5% of reading + 0.05 dS/m)|
|Precision||0.5% of BEC|
Relative Dielectric Permittivity
|Range||1 to 81|
Volumetric Water Content
|Range||0 to 100% (with M4 command)|
|Water Content Accuracy||
|Range||-50° to +70°C|
Note: The following shows notable compatibility information. It is not a comprehensive list of all compatible or incompatible products.
External RF sources can affect the probe’s operation. Therefore, the probe should be located away from significant sources of RF such as ac power lines and motors.
Multiple CS655 probes can be installed within 4 inches of each other when using the standard data logger SDI-12 “M” command. The SDI-12 “M” command allows only one probe to be enabled at a time.
The CS650G makes inserting soil-water sensors easier in dense or rocky soils. This tool can be hammered into the soil with force that might damage the sensor if the CS650G was not used. It makes pilot holes into which the rods of the sensors can then be inserted.
Current CS650 and CS655 firmware.
Note: The Device Configuration Utility and A200 Sensor-to-PC Interface are required to upload the included firmware to the sensor.
Number of FAQs related to CS655: 55
Expand AllCollapse All
No. The temperature sensor is located inside the sensor’s epoxy head next to one of the sensor rods. The stainless-steel rods are not thermally conductive, so the reported soil temperature reading is actually the temperature of the sensor head. If the CS650 or the CS655 is installed horizontally, which is the preferred method, then the sensor head will be at the same temperature as the soil, and the soil temperature value will be accurate. However, if the sensor is installed vertically, and/or with the sensor head above ground, the soil temperature reading will be less accurate. Because the sensor orientation is not known, no temperature correction was written into the firmware.
Yes. There is surge protection built into the sensor electronics. The sensor survives a surge of 2 kV at 42 ohm line-to-ground on digital I/O and 2 kV at 12 ohm line-to-ground on power. It also survives a surge of 2 kV at 2 ohm line-to-ground on the rods.
If additional surge protection is required, consider using the SVP100 Surge Voltage Protector DIN Rail with Mounting Hardware.
The CWS655 is a wireless sensor with measurement electronics, radio, and power supply all integrated in a single device. The CWS655, however, requires the use of a CWB100 base station radio connected to a data logger. Only the rods of the CWS655 should be buried in the soil; burying the body of the CWS655 will prevent the sensor from communicating with the CWB100.
The CS655 is a cabled multiparameter smart sensor that sends data by RS-232 serial or SDI-12 communication through a direct connection to a data logger. The CS655 is suitable for burial at any depth.
In soil that has a significant fraction of fines (loam, silt loam, silty clay loam, clay loam, clay), the CS655 is a suitable option because these soils tend to be more electrically conductive, and the CS655 operates over a larger range of electrical conductivity than the CS650. In applications where a smaller measurement volume is desired, such as larger greenhouse pots, the 12 cm long rods of the CS655 are preferable to the 30 cm long rods of the CS650.
If a system has multiple CS650 or CS655 sensors, it will be necessary to connect many wires to a 12 V supply and many wires to ground. The DIN-Rail Mounting Kit is useful for attaching many wires to the same source in a clean and organized way. For more details, see the 5458 DIN-Rail Terminal Kit instruction manual.
Other methods of connecting several wires together, such as terminal strips or wire nuts, would also work.
The CS650 and CS655 are warranted by Campbell Scientific to be free from defects in materials and workmanship under normal use and service for 12 months from the date of shipment. For further details, see the “Warranty” section of the CS650/CS655 manual.
The volumetric water content reading is the average water content over the length of the sensor’s rods.
A thermistor is encased in the epoxy head of the sensor next to one of the stainless-steel rods. This provides an accurate point measurement of temperature at the depth where that portion of the sensor head is in contact with the soil. This is why a horizontal placement is the recommended orientation of the CS650 or CS655. The temperature measurement is not averaged over the length of the sensor rods.
The bulk electrical conductivity (EC) measurement is made along the sensor rods, and it is an average reading of EC over that distance at whatever depth the rods are placed.