The 034A Windset (pn 10576) combined a 3-cup anemometer and vane into a single integrated package. It was constructed out of lightweight aluminum and designed for continuous long-term operation.
The "-L" on a product model indicates that the cable length is specified at the time of order.
Note: The following shows notable compatibility information. It is not a comprehensive list of all compatible or incompatible products.
The 034A can be attached to a CM202, CM204, or CM206 crossarm via a 17953 NU-RAIL fitting or a CM220 Right Angle Mounting Bracket. Alternatively, the 034A can be attached to the top of our stainless-steel tripods via the CM216 Sensor Mounting Kit. The CM216 extends 4 in. above the mast of a stainless-steel CM110, CM115, or CM120 tripod.
The 034A's anemometer uses one pulse count channel on the datalogger. Its wind vane requires one single-ended channel and access to an excitation channel (the excitation channel can be shared with other high impedance sensors).
The 034A's anemometer is measured by the PulseCount Instruction in CRBasic, and by Instruction 3 (Pulse Count) in Edlog. The wind vane is measured by the BrHalf Instruction in CRBasic and by Instruction 4 (Excite-Delay-SE) in Edlog. The measurements are typically processed for output with the Wind Vector instruction (not present in the CR500 or CR9000).
Number of FAQs related to 034A-L: 1