The ST-20 Argos Satellite Transmitter was well suited for remote meteorological and environmental data collection applications, including mobile and high-latitude installations. While Campbell Scientific's ST-20 Argos PTT was based on Telonics' ST-20, they were not equivalent; our ST-20 included a case, voltage regulator, and our SDC communication interface.Read More
Campbell Scientific's ST-20 was a Service Argos-certified Platform Transmitter Terminal (PTT ) that transferred data from the datalogger site to the Argos Data Collection and Location System via NOAA polar orbiting satellites.
Please note that when ordering the transmitter, the customer needed to provide Campbell Scientific with the number of buffers, repetition rate, decimal ID, Dex ID, frequency, and fail safe information.
While Campbell Scientific's ST-20 Argos PTT was based on Telonics' ST-20, they were not equivalent; our ST-20 included a case, voltage regulator, and our SDC communication interface.
|Supply Voltage||12 V, nominal|
|Output Impedance||50 ohm|
|High Power Level||800 mW typical; 27 to 31 dbm|
|Argos Transmission Frequency||401.618 to 401.680 MHz|
|Modulation (BPSK)||-1.1 to +1.1 ±0.1 radians|
|Current Drain at 12 Vdc||
|Spurious Emissions||-45 dB|
|Dimensions||19.05 x 5.72 x 3.18 cm (7.5 x 2.25 x 1.25 in.)|
|Weight||224 g (7.9 oz)|
Note: The following shows notable compatibility information. It is not a comprehensive list of all compatible or incompatible products.
The ST-20 requires a desiccated, non-condensing environment; a Campbell Scientific enclosure is recommended. Grommets and screws are provided to attach the ST-20's flanges to the backplate of our enclosures.