The SC-CPI adds CPI bus functionality to your CR800, CR850, CR1000, and CR3000 dataloggers, which is necessary for them to connect to the CDM (Campbell Distributed Module) devices and other CPI-enabled sensors. The SC-CPI connects to the datalogger using the C1, C2, and C3 ports.
The SC-CPI is not required for, or supported by, dataloggers with a native CPI interface. Campbell Scientific strongly recommends that CDMs be connected to dataloggers with a native CPI interface, such as the CR6.
|Current Drain||50 mA (active)|
|Dimensions||15.94 x 7.49 x 3.18 cm (6.28 x 2.95 x 1.25 in.) including ground lug|
|Weight||0.188 kg (0.415 lb)|
|Serial Connect to Datalogger||Screw terminal|
|Computer Configuration via Device Configuration Utility||USB|
Note: The following shows notable compatibility information. It is not a comprehensive list of all compatible or incompatible products.
A 5-conductor cable connects the datalogger to the SC-CPI, and an Ethernet cable connects the SC-CPI to the CDM.
The datalogger operating system must be version OS 26.
Execution of this download installs the SC-CPI Operating System on your computer for upload to the SC-CPI device with Device Configuration Utility.
Note: The SC-CPI OS V.02 or greater is required to use the CDM-VW300/305 with the CR3000/1000/800 datalogger that is using OS 28 or higher. The CDM-VW300/305 Operating System will also need to be upgraded to CDM-VW300/305 OS V.02. If using a CR3000/1000/800 datalogger with an older OS, an operating system upgrade to the datalogger will be required.
Number of FAQs related to SC-CPI: 2
To communicate with CDM devices using the CPI protocol, the CR3000, CR1000, CR850, and CR800 dataloggers require an SC-CPI device. Only one SC-CPI device per datalogger is required. It is expected that the design of future Campbell Scientific datalogger products will work with CPI-based modules directly.