The CS625 measures the volumetric water content from 0% to saturation. It is similar to our CS616 but is designed specifically for CR200(X) and CR300-series dataloggers. This reflectometer has a 0 to 3.3 V square wave frequency output that our CR300-series and CR200(X)-series dataloggers can measure.Read More
The CS625 consists of two 30-cm-long stainless steel rods connected to a printed circuit board. The circuit board is encapsulated in epoxy, and a shielded four-conductor cable is connected to the circuit board to supply power, enable probe, and monitor the output.
The CS625 measures the volumetric water content of porous media (such as soil) using the time-domain measurement method; a reflectometer (cable tester) such as the TDR100 is not required. This method consists of the CS625 generating an electromagnetic pulse. The elapsed travel time and pulse reflection are then measured and used to calculate soil volumetric water content.
The signal propagating along the parallel rods of the CS625 is attenuated by free ions in the soil solution and conductive constituents of the soil mineral fraction. In most applications, the attenuation is not enough to affect the CS625 response to changing water content, and the response is well described by the standard calibration. However, in soil with relatively high soil electrical conductivity levels, compacted soils, or soils with high clay content, the calibration should be adjusted for the specific medium. Guidance for making these adjustments is provided in the operating manual.
|Measurements Made||Volumetric water content of porous media (such as soil)|
|Water Content Accuracy||±2.5% VWC (using standard calibration with bulk EC of ≤ 0.5 dS m-1, bulk density of ≤ 1.55 g cm-3, and measurement range of 0% to 50% VWC)|
|Required Equipment||Measurement system|
|Soil Suitability||Long rods and lower frequency are well-suited for soft soil with low electrical conductivity (< 2 dS/m).|
|Operating Temperature Range||0° to 70°C|
|Probe-to-Probe Variability||±0.5% VWC in dry soil, ±1.5% VWC in typical saturated soil|
|Output||0 to 3.3 V square wave (with frequency dependent on water content)|
|Power Supply Voltage||5 Vdc minimum; 18 Vdc maximum|
|Enable Voltage||4 Vdc minimum; 18 Vdc maximum|
|Electromagnetic||CE compliant (Meets EN61326 requirements for protection against electrostatic discharge.)|
|Rod Spacing||32 mm (1.3 in.)|
|Rod Diameter||3.2 mm (0.13 in.)|
|Rod Length||300 mm (11.8 in.)|
|Probe Head Dimensions||85 x 63 x 18 mm (3.3 x 2.5 x 0.7 in.)|
|Cable Weight||35 g per m (0.38 oz per ft)|
|Weight||280 g (9.9 oz) without cable|
Note: The following shows notable compatibility information. It is not a comprehensive list of all compatible or incompatible products.
The RF emissions are below FCC and EU limits as specified in EN61326 if the CS625 is enabled less than 0.6 ms, and measurements are made less frequently than once a second. External RF sources can also affect the CS625 operation. Consequently, the CS625 should be located away from significant sources of RF such as ac power lines and motors.
The CS650G makes inserting soil-water sensors easier in dense or rocky soils. This tool can be hammered into the soil with force that might damage the sensor if the CS650G were not used. It makes pilot holes into which the rods of the sensors can then be inserted. It replaces both the 14383 and 14384.
Each CS625 requires a single-ended input channel. A control port is used to enable one or more probes.
Note: A maximum of four CS625 probes can be measured by one CR200(X) datalogger. Valid channel options are analog channels 1 through 4.
Number of FAQs related to CS625: 34
Expand AllCollapse All
300 m (1,000 ft).
Yes, as long as the data logger can detect a 0 to 3300 mV square wave over a frequency range of 29 to 67 kHz.
No. The output is too fast to be measured on the pulse channel of a 21X or CR7.
Fine roots do not significantly affect the CS616/CS625 reading.
Yes. The PeriodAvg() CRBasic instruction may be used to measure the CS625 on a CR1000, CR800-series, or CR3000 datalogger. To make this work, it is important that the Threshold parameter be set to 1650 mV as shown in the following example:
PortSet (1 ,1 ) ‘Enable CS625 by setting C1 high (orange wire to C1)
PeriodAvg (PA_uS,1,mV250,1,1650,0,100,10,1.0,0) ‘Read Period on SE1 (green wire to SE1)
PortSet (1 ,0) ‘Disable CS625 by setting C1 low
If a soil-specific calibration is performed, the CS616/CS625 may be used in soil with a maximum bulk electrical conductivity of 5 dS/m.
Yes. For program examples and guidance on using a multiplexer with one of these reflectometers, see the CS616 and CS625 instruction manual.
Yes. The dielectric permittivity of water varies with temperature, which will cause the CS616/CS625 period to decrease with increasing temperature and increase with decreasing temperature. In addition, the electrical conductivity of the soil water solution is temperature dependent, causing the CS616/CS625 period to increase with increasing temperature and decrease with decreasing temperature. The net effect of those two opposing forces depends on the soil texture and electrical conductivity. The temperature correction equation in the CS616 and CS625 instruction manual was developed using a sandy loam soil with relatively low electrical conductivity. For soils with finer texture or higher electrical conductivity, Campbell Scientific recommends a soil-specific temperature correction equation.
Running the cable through electrical conduit or PVC pipe will protect the cable from rodents. A trench 30 to 60 cm deep will protect it from most other human or animal activity. Some customers have found that extra cable can be coiled and left inside a box, such as an irrigation valve box or something similar. When using a box, seal any holes that are large enough for rodents to enter. When cables are exposed on the ground surface, some customers have found that wrapping the cables in the metal screening used for screen doors discourages animals from chewing on them.