The 4WPB100 provides completion resistors that allow your Campbell Scientific datalogger to measure a 100 ohm platinum resistive thermometer (PRT).Read More
|Dimensions||1.5 x 1.5 x 2.7 cm (0.6 x 0.6 x 1.0 in.) for body with prongs|
|Weight||6 g (0.2 oz)|
Current Limiting 10 kΩ Resistor
|Tolerance||±5% (@ 25°C)|
|Power Rating||0.25 W|
Completion Resistor (bulk metal foil)
|Resistance (Rf)||100 Ω|
|Tolerance||±0.01% (@ 25°C)|
|Maximum Temperature Coefficient||±0.8 ppm/°C|
|Power Rating||0.25 W (@ 70°C)|
The 4WPB100 uses two adjacent differential analog inputs; the first channel requires an adjacent analog ground channel to accept the "ground prong" of the 4WPB100. Not all datalogger terminal strips have this sequence on all channels, so please check your datalogger's wiring panel to confirm channel assignments, especially if you plan to use multiple Terminal Input Modules.
Newer CR10X Wiring Panels and the CR10XTCR Thermocouple Cover are deep enough to cover the datalogger input channels with a Terminal Input Module attached; older CR10TCRs are not deep enough.
When using CRBasic, the 4WPB100 is measured with the BrHalf4W Instruction followed by the PRT Instruction. When using Edlog, the 4WPB100 is measured with Instruction 9 (Full Bridge w/ Measured Compensation) followed by Instruction 16 (Temperature RTD).
|CR200(X) Series||CR800/CR850||CR1000||CR3000||CR9000X||CR6||CR300 Series|
*As shipped, the prongs on the 4WPB100 are too short to fit the CR7. If you're determined to use the 4WPB100 with a CR7, you can solder wire to the prongs, thereby extending their length.
Number of FAQs related to 4WPB100: 3
Expand AllCollapse All
Only one 4WPB100 is needed to measure the internal PRT in the radiometer.
The CR1000 requires a 4WPB100 to measure the internal PRT. (Dataloggers such as the CR3000 and CR5000 have the necessary PRT bridge module built in to measure the PRT.) Note that the CNR4-L also includes an internal thermistor, which can be directly measured by the CR1000. Because of this, when using a CR1000, Campbell Scientific typically recommends monitoring the internal temperature of the CNR4-L using its internal thermistor instead of the PRT.
The 4WPB100, 4WPB1K, and 4WHB10K terminal input modules can be used instead of the 3WHB10K module. However, to maintain the enhanced accuracy of the four-wire half-bridge instruction, it is pertinent when adding the fourth wire to maintain the same resistance as the other three wires (same wire size and manufacturer). Note that simply adding a short jumper wire on the 4WPB100, 4WPB1K, or 4WHB10K modules to complete the fourth wire will not maintain the accuracy.