The CS305-ETM pyranometer, manufactured by Apogee, is similar to the CS300 but uses a different fixed multiplier and includes a leveling mount.Read More
The "-ET" version of this product has a cable that terminates in a connector (instead of pigtails) and is the ideal length for attachment to an ET106 or ET107 Weather Station.
The CS305-ETM measures total sun and sky solar radiation using a silicon photocell. The spectral range is 300 to 1000 nanometers, which encompasses most of the shortwave radiation that reaches the Earth's surface.
The standard output is 0.005 mV per W m-2, which provides a signal of 5 mV in full sunlight (1000 W m-2). All our data loggers, except the CR200(X) series, can measure this output. Therefore both the ET107 and ET106 are compatible.
|Light Spectrum Waveband||300 to 1100 nm|
|Measurement Range||0 to 2000 W m-2 (full sunlight ≈ 1000 W m-2)|
|Absolute Accuracy||±5% for daily total radiation|
|Temperature Response||< 1% (at 5° to 40°C)|
|Long-Term Stability||< 2% per year|
|Diameter||2.4 cm (0.9 in.)|
|Cable Length||152.4 cm (60 in.)|
|Height||2.5 cm (1.0 in.)|
|Weight||65 g (2.3 oz)|
The connector attaches to an ET107 or ET106 Weather Station.
Number of FAQs related to CS305-ETM: 3
Expand AllCollapse All
Most Campbell Scientific sensors are available as an –L, which indicates a user-specified cable length. If a sensor is listed as an –LX model (where “X” is some other character), that sensor’s cable has a user-specified length, but it terminates with a specific connector for a unique system:
If a sensor does not have an –L or other –LX designation after the main model number, the sensor has a set cable length. The cable length is listed at the end of the Description field in the product’s Ordering information. For example, the 034B-ET model has a description of “Met One Wind Set for ET Station, 67 inch Cable.” Products with a set cable length terminate, as a default, with pigtails.
If a cable terminates with a special connector for a unique system, the end of the model number designates which system. For example, the 034B-ET model designates the sensor as a 034B for an ET107 system.
Not every sensor has different cable termination options. The options available for a particular sensor can be checked by looking in two places in the Ordering information area of the sensor product page:
If a sensor is offered in an –ET, –ETM, –LC, –LQ, or –QD version, that option’s availability is reflected in the sensor model number. For example, the 034B is offered as the 034B-ET, 034B-ETM, 034B-LC, 034B-LQ, and 034B-QD.
All of the other cable termination options, if available, are listed on the Ordering information area of the sensor product page under “Cable Termination Options.” For example, the 034B-L Wind Set is offered with the –CWS, –PT, and –PW options, as shown in the Ordering information area of the 034B-L product page.
Note: As newer products are added to our inventory, typically, we will list multiple cable termination options under a single sensor model rather than creating multiple model numbers. For example, the HC2S3-L has a –C cable termination option for connecting it to a CS110 instead of offering an HC2S3-LC model.
Many Campbell Scientific sensors are available with different cable termination options. These options include the following:
Note: The availability of cable termination options varies by sensor. For example, sensors may have none, two, or several options to choose from. If a desired option is not listed for a specific sensor, contact Campbell Scientific for assistance.