Image Not Available
To ensure the optimum operation of the Strike Guard System, a basic factory service is recommended every three to four years from the time of purchase or the last factory service. More frequent service is recommended for high humidity and high salt spray (coastal) environments.
Basic factory service for the Strike Guard Lightning Sensor includes the following: replacement of the internal lithium batteries, replacement of the desiccant, a new fiber-optic transmitter, a new optical diffuser, and firmware updates to the latest factory standard.
An RMA (Returned Material Authorization) is required.