|Measurement Description||Wind direction|
|Measurement||Analog potentiometer (half bridge)|
|Range||0° to 360°|
|Operating Temperature Range||-55° to +60°C|
|Starting Threshold||1.0 m/s (2.2 mph)|
|Swept Diameter||27 cm (10.5 in.)|
|Length||21 cm (8.3 in.)|
|Height||12 cm (4.3 in.)|
|Weight||0.14 kg (0.3 lb)|
Note: The following shows notable compatibility information. It is not a comprehensive list of all compatible or incompatible products.
The Wind Vane requires one single-ended channel and access to an excitation channel (the excitation channel can be shared with other high impedance sensors).
The wind vane is measured by the BrHalf Instruction in CRBasic and by Instruction 4 (Excite-Delay-SE) in Edlog. The measurements are typically processed for output with the Wind Vector Instruction (not present in the CR500 or CR9000).
Number of FAQs related to 200P: 1