The 27450 is molecular sieve desiccant material to refill the bottles in the 27423 Scrub Module. This desiccant is included with the scrub module and is available as a replacement part. The 27450 consists of 13 x 1.6-2.5 mm beads in a 250 g bottle.
|Amount||250 g (8.82 oz)|
Number of FAQs related to 27450: 4
Expand AllCollapse All
The molecular sieve is a non-hazardous material that can be shipped to any country.
The molecular sieve is a direct replacement for the old magnesium perchlorate bottles. The molecular sieve may be used for any Campbell Scientific analyzer that used the old bottles.
The bottles of sieve for drop-in replacement contain the pellets and a membrane on top. The membrane is necessary to keep the pellets contained while allowing gas to pass over the zeolite. The bottle has the same footprint as the old magnesium perchlorate bottles. The amount in each bottle is listed on the bottle. The amount of sieve needed for each analyzer is the following:
The molecular sieve has been demonstrated here by our engineering department to be effective at removing CO2 and H2O from the air sample. The change was made for two reasons: