Communication through the data logger’s CS I/O port requires an interface that converts the computer’s RS-232 voltage levels to the CMOS levels of the data logger. Typically, that interface is an SC32B. The SC32B connects to the data logger’s CS I/O port via an SC12 or serial cable and connects to a computer’s 9-pin serial port via a serial cable. When the computer does not have a serial port, a 9-pin serial-to-USB cable, such as pn 17394, can be used with the SC32B. Alternatively, instead of the SC32B, use the SC-USB.