The CS300-QD, manufactured by Apogee, measures total sun and sky solar radiation for solar, agricultural, meteorological, and hydrological applications. Its spectral range of 360 to 1120 nanometers encompasses most of the short-wave radiation that reaches the Earth's surface. The CS300-QD supports continuous outdoor use.Read More
The "-QD" version of this product has a cable that terminates in a military style connector (instead of pigtails) and is the ideal length for attachment to a RAWS-F Fire Weather Quick Deployment Station.
With this version of the CS300 pyranometer, the cable has a military style connector instead of pigtails for attachment to the RAWS-F Fire Weather Quick Deployment Station.
The CS300-QD uses a silicon photovoltaic detector mounted in a cosine-corrected head to provide solar radiation measurements. Its dome-shaped head prevents water from accumulating on the sensor head. To eliminate internal condensation, the sensor head is potted solid and the cable is shielded with a rugged Santoprene casing. The CS300 is calibrated against a Kipp & Zonen CM21 thermopile pyranometer to accurately measure sun plus sky radiation.
|Light Spectrum Waveband||360 to 1120 nm (wavelengths where response is 10% of maximum)|
|Measurement Range||0 to 2000 W m-2 (full sunlight ≈1000 W m-2)|
|Absolute Accuracy||±5% for daily total radiation|
|Sensitivity||5 W m-2 mV-1 (0.2 mV W-1 m-2)|
|Cosine Correction Error||±5% at 75° zenith angle; ±2% at 45° zenith angle|
|Temperature Response||0.04 ± 0.04% per °C|
|Response Time||< 1 ms|
|Long-Term Stability||< 2% per year|
|Operating Temperature Range||-40° to +70°C|
|Relative Humidity Range||0 to 100%|
|Output||0.2 mV per W m-2|
|Cable Length||2.44 m (8 ft)|
|Diameter||2.4 cm (0.9 in.)|
|Height||2.5 cm (1.0 in.)|
|Weight||65 g (2.3 oz) with 2-m (6.6-ft) lead wire|
The military style connector attaches to the RAWS-F Fire Weather Quick Deployment Station, which has an on-board CR1000.
Accurate measurements require the sensor to be leveled using a 18356 leveling fixture. This leveling fixture incorporates a bubble level and three adjusting screws. The 18356 mounts to a crossarm or a tripod or tower mast using the CM225 mounting stand. The CS300 should be mounted away from all obstructions and reflective surfaces that might adversely effect the measurement.
One differential analog input channel per probe is required.
Number of FAQs related to CS300-QD: 3
Expand AllCollapse All
Not every sensor has different cable termination options. The options available for a particular sensor can be checked by looking in two places in the Ordering information area of the sensor product page:
If a sensor is offered in an –ET, –ETM, –LC, –LQ, or –QD version, that option’s availability is reflected in the sensor model number. For example, the 034B is offered as the 034B-ET, 034B-ETM, 034B-LC, 034B-LQ, and 034B-QD.
All of the other cable termination options, if available, are listed on the Ordering information area of the sensor product page under “Cable Termination Options.” For example, the 034B-L Wind Set is offered with the –CWS, –PT, and –PW options, as shown in the Ordering information area of the 034B-L product page.
Note: As newer products are added to our inventory, typically, we will list multiple cable termination options under a single sensor model rather than creating multiple model numbers. For example, the HC2S3-L has a –C cable termination option for connecting it to a CS110 instead of offering an HC2S3-LC model.
Most Campbell Scientific sensors are available as an –L, which indicates a user-specified cable length. If a sensor is listed as an –LX model (where “X” is some other character), that sensor’s cable has a user-specified length, but it terminates with a specific connector for a unique system:
If a sensor does not have an –L or other –LX designation after the main model number, the sensor has a set cable length. The cable length is listed at the end of the Description field in the product’s Ordering information. For example, the 034B-ET model has a description of “Met One Wind Set for ET Station, 67 inch Cable.” Products with a set cable length terminate, as a default, with pigtails.
If a cable terminates with a special connector for a unique system, the end of the model number designates which system. For example, the 034B-ET model designates the sensor as a 034B for an ET107 system.
Many Campbell Scientific sensors are available with different cable termination options. These options include the following:
Note: The availability of cable termination options varies by sensor. For example, sensors may have none, two, or several options to choose from. If a desired option is not listed for a specific sensor, contact Campbell Scientific for assistance.