The CC5MPXWD is similar to the CC5MPX, except it includes an internal defroster that prevents and removes light frost and icing from the camera's window. This high-resolution digital camera produces JPG images with a resolution of up to 5 megapixels, and shoots videos with a resolution of up to 720P. It has an integrated environmentally sealed enclosure that protects the camera from moisture and high humidity. An SD memory card for image storage is available as an option.Read More
The CC5MPXWD can function as a stand-alone camera or can be connected to a Campbell Scientific data logger or another PakBus device. It has several options for image acquisition and storage. You can take still images or videos based on an internal timer, motion detection, or a trigger from an external PakBus device (such as a data logger). The images or videos can then be stored on an SD card, saved in a data logger’s memory, sent to a PC via email, or transmitted to an FTP server.
The CC5MPXWD's defroster consists of an aluminum ring with a resistive type heating element. Springs are used to press the defroster against the camera's window.
The defroster turns on when the internal temperature is below a specified value and not in the low powered state. You can also specify the number of minutes before a self-timed schedule that the defroster will turn on.
With extreme icing or riming conditions and low temperatures, the window defroster may not be able to clear the window due to the power constraints of the defroster.
|Heating Element Resistance||18 Ω|
9 to 16 Vdc
Operating at voltages greater than 16 Vdc can damage the camera.
|Operating Temperature Range||-40° to +60°C|
|Clock Accuracy||±2 minutes per year (-40° to +60°C)|
|Photo or Video Capture Triggers||Two independent self timers, external trigger, motion detection, web page control|
|Programmable Still Image Resolutions||
Capable of up to 720P for the following:
|Communication Interfaces||RS-232 port; RS-485 port; Ethernet 10/100|
|Communications Protocols||PakBus, FTP, email, web page interface via web browser|
|Maximum Baud Rate||115.2 kbps (for RS-232 and RS-485 only)|
|Low-Light Capability||Not designed for low-light conditions.|
|Diameter||9.3 cm (3.7 in.)|
|Length||22 cm (8.7 in.)|
|Weight||1.06 kg (2.34 lb)|
Communication Switched Power Output
|Maximum Output Current||750 mA|
Current Drain (camera only)
|Maximum||250 mA (at 12 Vdc)|
|Quiescent||≤1 mA (off power mode)|
Current Drain (with defroster on)
|Typical||1.0 A (@ 12 Vdc)|
|Maximum||1.2 A (@ 16 Vdc)|
|Standard Lens||4 to 12 mm, 27° to 80° FOV|
External Input Signal
|Logic Low Level||0.65 Vdc (-20 Vdc absolute minimum)|
|Logic High Level||> 2.0 Vdc (+20 Vdc absolute maximum)|
|Minimum Pulse Width||10 ms|
Memory Card Interface
|Type||Secure Digital (SD)|
|File Type||JPEG (image), AVI (video)|
|Size||16 GB or less|
Photo Capture Time
|-NOTE-||Time is from wake-up to start of capture.|
|Fully On||< 1 s (5 MP images take longer.)|
|Partially On||10 s|
|Deep Sleep||15 s|
|Off State||90 s|
Note: The following shows notable compatibility information. It is not a comprehensive list of all compatible or incompatible products.
|CR10X (retired)||The CR10X must be the -2M version and must use the PakBus operating system.|
|CR23X (retired)||The CR23X must be the -2M version and must use the PakBus operating system.|
|CR510 (retired)||The CR510 must be the -2M version and must use the PakBus operating system.|
The 18549 Camera Mounting Kit is used to mount the CC5MPXWD camera to a CM202, CM204, or CM206 crossarm. The crossarm then attaches to a tripod, tower, or user-supplied pole (1.0-in. to 2.1-in. outer diameter). The bracket’s design facilitates the aiming of the camera.
The CC5MPXWD is compatible with any PakBus data logger that has 2 MB or greater of on-board memory. The data logger must have a current operating system.
The CC5MPXCBL1 Power and I/O Cable connects the camera to:
An Ethernet cable is required to configure the camera. Typically, the CC5MPXCBL2 Environmental Cable is used. The Ethernet cable connects the camera to a network router, cellular modem, or laptop. With this connection, the user can access a web interface that:
Current CC5MPX Camera Operating System.
Note: Do not extract the firmware update file “CC5MPX_FW_1_25_1.tar.gz”.This compressed file gets sent to the CC5MPX as is.View Update History
Current firmware for the CC5MPX Peripheral Interface Controller (PIC)
Note: In order for the CC5MPX to accept the PIC firmware the CC5MPX OS 1.23 or higher must be running in the camera. Retrieve the CC5MPX_PIC_UPDATE_V2_JUNE_17_2014.hex.gz firmware from the .zip file. Do not extract the firmware update file "CC5MPX_PIC_UPDATE_V2_JUNE_17_2014.hex.gz". This compressed file must be sent to the CC5MPX as a “.gz” file. Details related to updating firmware can be found in the CC5MPX manual.
Number of FAQs related to CC5MPXWD: 14
Expand AllCollapse All
This depends. Typically, most camera applications require an environmentally sealed cable, such as the CC5MPXCBL2-L, rather than a generic Ethernet cable. However, for indoor or temporary applications where the cable will not be exposed to weather elements, use a standard 10Base-T CAT5e Ethernet cable, such as pn 28900.
The CC5MPXCBL2-L is an environmentally sealed Ethernet (RJ45) cable that provides access to the network functionalities of the CC5MPX family of cameras. This cable is used with a camera to do the following: send emails, transfer images via FTP (accessing either a web server or an optional internal SD memory card), synchronize with a network time server, or gain access to the camera’s configuration web page. The CC5MPXCBL2-L can also be used to connect the CC5MPX directly to an Ethernet port, such as a cellular modem or wireless router. In addition, Campbell Scientific recommends using this cable if the CC5MPX is mounted on a tall tower or in a location that is not easily accessible.
Note: The camera’s internally served configuration web page is the only way to view the real-time streaming video that is used to manually zoom and focus the camera.
Although the CC5MPX’s enclosure is weather resistant, it is not designed for underwater use. However, if a watertight enclosure is rated suitable for saltwater environments, and there is physical space for the CC5MPX within the enclosure, then the camera will function as designed.
Note: Campbell Scientific does not offer any watertight enclosures rated suitable for saltwater environments.
The CC5MPX uses a standard C-Type mount lens. If a lens fits within the lens tube barrel, or if the camera will be used in an environment where the front lens tube barrel is unnecessary, other lenses, such as a zoom or fisheye may be used.
Note: Use of an alternate lens may void the product warranty.
This is done by sending images via PakBus to a Campbell Scientific data logger with a cellular modem telemetry option. The cable that is used is the CC5MPXCBL1-L Power and I/O Cable, which provides power from the data logger, as well as an RS-232 or RS-485 connection.
PoE, or Power over Ethernet, is commonly used as a way to power devices and equipment using the unused conductors in a standard Ethernet cable. To accomplish this, there must be a sending device, or power source equipment, as well as a powered device.
The CC5MPX family of cameras does not work with PoE. For the family of CC5MPX cameras to be powered, a separate accessory, the CC5MPXCBL1-L cable, must be used to provide both power and RS-232/RS-485 communications.
No. A data logger is not necessary if the CC5MPX will be used in a stand-alone mode. In essence, a camera that is used without a data logger is an IP web camera. The camera does, however, need access to a 9 to 30 Vdc power supply.
Images can be obtained from the CC5MPX’s Ethernet connection through either FTP or email.
This may be possible. Contact Campbell Scientific for assistance with this.