The A100LK is an anemometer suitable for your general meteorology applications and wind-energy surveys. Its low power consumption and wide power-supply range make it popular for remote locations with no access to ac power, and its pulse/frequency signal is ideal for use with Campbell Scientific dataloggers. The sensor is constructed from anodized aluminum alloys, stainless steel, and weather-resistant plastics for long life. It is used as part of our WMS100 wind-monitoring system.Read More
Rotation of the A100LK’s three-cup rotor is electronically converted to pulse output signals proportional to wind speed. The A100LK produces a higher rate of pulses per revolution (up to 13) compared to relay-based sensors, making it suitable for wind surveying where turbulence needs to be estimated.
|Threshold||0.15 ms-1 (starting speed 0.2 ms-1, stopping speed 0.1 ms-1)|
|Maximum Speed||77.22 ms-1|
|Accuracy||1% ±0.1 ms-1|
|Distance Constant||2.3 m ±10%|
Supplied for anemometer and rotor at one test speed to an accuracy of ±1% at +15°C.
(12 Vdc supply and an analog output load of 1 MΩ)
|Operating Temperature Range||-30° to +70°C|
|Rotor||15.2 cm (6 in.) diameter three-cup rotor|
|Supply Voltage||6.5 to 28 Vdc|
2 mA maximum
1.6 mA typical (no output load)
|Power-up Time||5 s|
|Surge Protection||Vector PC3L2 anti-surge module fitted|
|Housing Diameter||5.5 cm (2.2 in.)|
|Height||19.5 cm (7.68 in.)|
|Weight||490 g (17.3 oz) including 3 m (10 ft) cable|
The A100LK uses one pulse count channel on the datalogger.
The A100LK is read by the PulseCount Instruction in CRBasic and by Instruction 3 (Pulse Count) in Edlog. Wind speed measurements can be taken and processed with any of the dataloggers.
|CR200(X) Series||CR800/CR850||CR1000||CR3000||CR9000X||CR6 Series||CR300 Series|
Number of FAQs related to A100LK-L: 6
Expand AllCollapse All
No. Cables are attached and sealed on the sensor by the manufacturer. Whenever possible, purchase a sensor with the desired cable length. Alternatively, a new sensor can be purchased with a foot of cable and then spliced onto the existing cable. Splicing cable together, however, increases the likelihood that water may enter the cable and cause shorting, corrosion, and some other potential issues, which in turn can cause measurement issues. Because of the potential issues, do not splice any sensor cable without first contacting Campbell Scientific to discuss the sensor in detail.
Yes. Every A100LK-L ships with a MEASNET calibration certificate.
Not every sensor has different cable termination options. The options available for a particular sensor can be checked by looking in two places in the Ordering information area of the sensor product page:
If a sensor is offered in an –ET, –ETM, –LC, –LQ, or –QD version, that option’s availability is reflected in the sensor model number. For example, the 034B is offered as the 034B-ET, 034B-ETM, 034B-LC, 034B-LQ, and 034B-QD.
All of the other cable termination options, if available, are listed on the Ordering information area of the sensor product page under “Cable Termination Options.” For example, the 034B-L Wind Set is offered with the –CWS, –PT, and –PW options, as shown in the Ordering information area of the 034B-L product page.
Note: As newer products are added to our inventory, typically, we will list multiple cable termination options under a single sensor model rather than creating multiple model numbers. For example, the HC2S3-L has a –C cable termination option for connecting it to a CS110 instead of offering an HC2S3-LC model.
Most Campbell Scientific sensors are available as an –L, which indicates a user-specified cable length. If a sensor is listed as an –LX model (where “X” is some other character), that sensor’s cable has a user-specified length, but it terminates with a specific connector for a unique system:
If a sensor does not have an –L or other –LX designation after the main model number, the sensor has a set cable length. The cable length is listed at the end of the Description field in the product’s Ordering information. For example, the 034B-ET model has a description of “Met One Wind Set for ET Station, 67 inch Cable.” Products with a set cable length terminate, as a default, with pigtails.
If a cable terminates with a special connector for a unique system, the end of the model number designates which system. For example, the 034B-ET model designates the sensor as a 034B for an ET107 system.
Many Campbell Scientific sensors are available with different cable termination options. These options include the following:
Note: The availability of cable termination options varies by sensor. For example, sensors may have none, two, or several options to choose from. If a desired option is not listed for a specific sensor, contact Campbell Scientific for assistance.
It is possible that an older version of Short Cut is being used. Download the latest version of Short Cut.
If the latest version of Short Cut has already been downloaded, open the program.