The HMP45C-LQ is a rugged, accurate temperature and relative humidity probe. The sensor measures relative humidity over the 0 to 100% RH range and temperature over the -40° to +60°C range.Read More
The "-LQ" version of this product terminates in a military style connector (instead of pigtails) for use with a RAWS-P Permanent Remote Automated Weather Station. The length of the cable is specified at the time of order.
For this version of the HMP45C, the cable terminates in a military style connector and has a user-specified lead length. The connector attaches to the RAWS-P enclosure, which mounts to a CM6/CM10 tripod, CM110-series tripod, or UT-series tower. The probe uses a HUMICAP capacitive polymer H chip to measure relative humidity over the 0 to 100% RH range. A PRT measures temperature over the -40° to +60°C range.
|Supply Voltage||12 Vdc nominal (typically powered by datalogger)|
|Current Consumption||≤ 4 mA (active)|
|Temperature Sensor||1000 ohm Platinum Resistance Thermometer|
|Measurement Range||-39.2° to +60°C|
|Output Signal Range||0.008 to 1.0 V|
|Accuracy||See graph in HMP45C product brochure.|
|Sensor||Vaisala’s HUMICAP H-chip|
|Measurement Range||0.8% to 100% RH (non-condensing)|
|Output Signal Range||0.008 to 1 Vdc|
|Accuracy at 20°C||
|Temperature Dependence||±0.05% RH/°C|
|Long-Term Stability||Typically, better than 1% RH per year|
|Response Time||15 seconds with membrane filter (at 20°C, 90% response)|
|Settling Time||500 milliseconds|
The sensor cable terminates in a military style connector and a user-specified lead length. The connector attaches to the RAWS-P enclosure.
The HMP45C-LQ should be housed in a 41003-5 10-plate naturally-aspirated solar radiation shield. This radiation shield mounts to a tripod or tower mast, crossarm, or user-supplied pole.