Hi, I would like some help setting up a program or the settings in LoggerNet for Device config. I am looking to run all channels possible for Strain Gauge, along with an AM 16/32B but for now I only need one (CH. 5) to work well. The strain gauges are 350 OHM GF 2.07, they are quarter bridge as well. I set them up on a board with resistors to complete the wheatstone bridge, so they should function as full-bridge. Thank you!
That type of sensor is supported in the Short Cut program generator. You need to add the AM16/32 to the CR1000 from the Device category. Then, add the sensors to the AM16/32. Quarter Bridge Strain is a sensor in the Geotechnical & Structural category.
ShortCut will create the program and wiring diagram for you.
This post is under review.