The CS616-LC is a version of the CS616 that has a cable terminated in a connector rather than the standard pigtails. The connector can be attached to the ET107, ET106, or MetData1 weather stations.
The CS616-LC measures the volumetric water content from 0% to saturation. It uses time-domain measurement methods, but no reflectometer (cable tester) such as the TDR100 is required. The probe rods can be inserted from the surface or the probe can be buried at any orientation to the surface. It is rugged and designed for long-term, unattended use.Read More
The CS616-LC is a version of the CS616 that has a cable terminated in a connector that attaches to the ET107, ET106, or MetData1 weather stations.
The CS616-LC measures the volumetric water content from 0% to saturation. It uses time-domain measurement methods. The probe rods can be inserted from the surface or the probe can be buried at any orientation to the surface. It is rugged, and is designed for long-term, unattended use.
|Output||±0.7 V square wave with frequency dependent on water content|
|Accuracy||±2.5% VWC using standard calibration with bulk electrical conductivity of ≤ 0.5 dS m-1, bulk density of ≤ 1.55 g cm-3, and measurement range of 0 to 50% VWC|
|Power Supply Voltage||
|Rod Spacing||32 mm (1.3 in.)|
|Rod Diameter||3.2 mm (0.13 in.)|
|Rod Length||300 mm (11.8 in.)|
|Probe Head Dimensions||85 x 63 x 18 mm (3.3 x 2.5 x 0.7 in.)|
|Weight||280 g (9.9 oz) without cable|
|Cable Weight||35 g per m (0.38 oz per ft)|
Number of FAQs related to CS616-LC: 3
Expand AllCollapse All
Not every sensor has different cable termination options. The options available for a particular sensor can be checked by looking in two places in the Ordering information area of the sensor product page:
If a sensor is offered in an –ET, –ETM, –LC, –LQ, or –QD version, that option’s availability is reflected in the sensor model number. For example, the 034B is offered as the 034B-ET, 034B-ETM, 034B-LC, 034B-LQ, and 034B-QD.
All of the other cable termination options, if available, are listed on the Ordering information area of the sensor product page under “Cable Termination Options.” For example, the 034B-L Wind Set is offered with the –CWS, –PT, and –PW options, as shown in the Ordering information area of the 034B-L product page.
Note: As newer products are added to our inventory, typically, we will list multiple cable termination options under a single sensor model rather than creating multiple model numbers. For example, the HC2S3-L has a –C cable termination option for connecting it to a CS110 instead of offering an HC2S3-LC model.
Most Campbell Scientific sensors are available as an –L, which indicates a user-specified cable length. If a sensor is listed as an –LX model (where “X” is some other character), that sensor’s cable has a user-specified length, but it terminates with a specific connector for a unique system:
If a sensor does not have an –L or other –LX designation after the main model number, the sensor has a set cable length. The cable length is listed at the end of the Description field in the product’s Ordering information. For example, the 034B-ET model has a description of “Met One Wind Set for ET Station, 67 inch Cable.” Products with a set cable length terminate, as a default, with pigtails.
If a cable terminates with a special connector for a unique system, the end of the model number designates which system. For example, the 034B-ET model designates the sensor as a 034B for an ET107 system.
Many Campbell Scientific sensors are available with different cable termination options. These options include the following:
Note: The availability of cable termination options varies by sensor. For example, sensors may have none, two, or several options to choose from. If a desired option is not listed for a specific sensor, contact Campbell Scientific for assistance.