The HMP60, manufactured by Vaisala, probe measures temperature for the range of -40° to 60°C, and relative humidity for the range of 0 to 100% RH. It is suitable for long-term, unattended monitoring, and is compatible with all Campbell Scientific dataloggers.Read More
The HMP60 uses the INTERCAP capacitive RH chip. This chip is field-replaceable, which eliminates the downtime typically required for the recalibration process.
|Supply Voltage||5 to 28 Vdc (typically powered by datalogger’s 12 V supply)|
|Filter Description||0.2 µm Teflon membrane|
|Settling Time||1 s|
|Sensor Diameter||1.2 cm (0.5 in.)|
|Filter Diameter||1.2 cm (0.5 in.)|
|Length||7.1 cm (2.8 in.)|
|Weight||0.05 kg (0.1 lb) with 1.83-m (6-ft) cable|
|Body Material||AISI 316 stainless steel|
|Filter Cap Material||Chrome-coated ABS plastic|
|Temperature Sensor||1000 ohm Platinum Resistance Thermometer|
|Measurement Range||-40° to +60°C|
|Sensor||Vaisala’s INTERCAP capacitive chip|
|Measurement Range||0 to 100% RH (non-condensing)|
Relative Humidity - Accuracy
|Typical Accuracy at -40° to 0°C||
|Typical Accuracy at 0° to 40°C||
|Typical Accuracy at 40° to 60°C||
The HMP60 is typically housed in a 41303-5A 6-plate naturally aspirated shield.
This probe may also be housed in a 41003-5 10-plate shield if additional hardware is used. A 41322 Adapter Plate allows the HMP60 to be mounted in the lower part of the 41003-5 10-plate shield. Alternatively, a 41381 extension tube and the 6637 split nut plug can be used to mount the HMP60 in a higher part of the 41003-5; the pn 41381 cable is also required.
Both the 41303-5A and the 41003-5 attach to a crossarm, mast, or user-supplied pipe with a 1.0-in. to 2.1-in. outer diameter.
|CR200(X) Series||CR800/CR850||CR1000||CR3000||CR6||CR9000X||CR300 Series|
Number of FAQs related to HMP60-L: 8
Expand AllCollapse All
The HMP60-L (with the Campbell Scientific cable) cannot be used with the 41003-5 10-Plate Radiation Shield without the use of additional hardware. For more information, refer to the Compatibility information on the HMP60-L product page. The standard radiation shield for the HMP60-L is the 41303-5A.
In general, the element should be replaced every two years. By doing a field check (that is, comparing the RH reading with a reference), the element can be used until the relative humidity measurement is out of tolerance.
The most significant difference is that the CS215-L has an SDI-12 output and that the HMP60-L has a voltage output. Additional differences are outlined in the “Air Temperature & Relative Humidity” brochure, which compares all of the Campbell Scientific temperature and relative humidity sensors.
Not every sensor has different cable termination options. The options available for a particular sensor can be checked by looking in two places in the Ordering information area of the sensor product page:
If a sensor is offered in an –ET, –ETM, –LC, –LQ, or –QD version, that option’s availability is reflected in the sensor model number. For example, the 034B is offered as the 034B-ET, 034B-ETM, 034B-LC, 034B-LQ, and 034B-QD.
All of the other cable termination options, if available, are listed on the Ordering information area of the sensor product page under “Cable Termination Options.” For example, the 034B-L Wind Set is offered with the –CWS, –PT, and –PW options, as shown in the Ordering information area of the 034B-L product page.
Note: As newer products are added to our inventory, typically, we will list multiple cable termination options under a single sensor model rather than creating multiple model numbers. For example, the HC2S3-L has a –C cable termination option for connecting it to a CS110 instead of offering an HC2S3-LC model.
Most Campbell Scientific sensors are available as an –L, which indicates a user-specified cable length. If a sensor is listed as an –LX model (where “X” is some other character), that sensor’s cable has a user-specified length, but it terminates with a specific connector for a unique system:
If a sensor does not have an –L or other –LX designation after the main model number, the sensor has a set cable length. The cable length is listed at the end of the Description field in the product’s Ordering information. For example, the 034B-ET model has a description of “Met One Wind Set for ET Station, 67 inch Cable.” Products with a set cable length terminate, as a default, with pigtails.
If a cable terminates with a special connector for a unique system, the end of the model number designates which system. For example, the 034B-ET model designates the sensor as a 034B for an ET107 system.
Note the difference between calibration and a field check. Calibration cannot be done in the field, as it requires an experienced technician and specialized equipment.
Field checks of measurements can be done to determine if the data make sense with the real-world conditions. Follow these steps to field check a sensor:
Many Campbell Scientific sensors are available with different cable termination options. These options include the following:
Note: The availability of cable termination options varies by sensor. For example, sensors may have none, two, or several options to choose from. If a desired option is not listed for a specific sensor, contact Campbell Scientific for assistance.
To incorporate a sensor that is compatible with wireless sensor interfaces into a wireless network, a CWS900-series wireless sensor interface is needed, as well as an A205 CWS-to-PC interface to configure it.